IntelliPAP BiLevel Machine with SmartCode
Save $37.95 on the ComfortLine Heated Tubing when you buy the IntelliPAP. Click the "Customize & Buy" button below to view more discounts.
The IntelliPAP BiLevel is a versatile therapy solution for users with high pressure needs. Take advantage of adjustable IPAP and EPAP settings along with advanced reporting. Now with SmartCode, the IntelliPAP can generate therapy reports through the Internet. This unit is backed by a 5 Year industry-leading warranty and one of the quietest machines on the market.
Manufactured by DeVilbiss.
15% of BiLevel Machine product buyers choose this product.
4.8 of 5 16 Reviews
Have a question? Call us!1-800-356-5221(Mon-Fri 8AM-10PM CST, Sat 8AM-6PM CST)
DeVilbiss offers free firmware upgrades for the IntelliPAP Series of CPAP machines. For example, The AutoAdjust update allows existing owners to add SmartCode and SmartFlex technologies. You can check for the most recent firmware update here: Model DV5 Series Firmware Updates.
Once you have downloaded the firmware software, use the IntelliPAP CPAP to PC Firmware Upgrade Serial Cable to upload your DeVilbiss firmware to the machine from your computer. The cable connects to the back of the CPAP and to any PC serial port on your computer.
IntelliPAP users must have a PC serial port available for the firmware upgrade to work. The DeVilbiss software only supports a serial connection; USB-to-USB or other connections will not work. If your computer does not come with a serial port, CPAP.com provides a USB male to Serial male (9-pin) DB-9 RS-232 adapter that allows users to update their IntelliPAP CPAP machines through your computer's USB port. If you do not have a serial port available, then you will need to buy both cords together: IntelliPAP Firmware Upgrade Cable with USB-to-Serial PC Adapter.
Here are some basic instructions on how to install a firmware update once you have downloaded the software to your PC:
2. Connect the CPAP to a free serial port with cable DV51D-615 and apply AC power to the CPAP device.
3. Close your web browser and all other open applications.
4. Run the DV5xFirmware.exe application Note: No files will be permanently installed on your computer.
5. The software will automatically search the computer serial ports for a supported CPAP device. If found, the "Connected CPAP Details" section will show information about the attached device.
6. Click the "Start" button to begin the firmware update. The progress bar and status messages will be updated as the firmware update proceeds. Wait for the update to complete before using your computer. Do not run any other applications during the update.
7. When finished, the software shows the results on the screen (OK or Fail). If the update was successfully completed, you can close the program by clicking the "Exit" button. If the update fails press "Start" to try again.
On-Board Memory and SD Card Requirements
The IntelliPAP Bilevel can store up to 90 days of therapy data, and up to 3 years of usage data. If an SD memory card is used to retrieve data from the machine it must be a file format of FAT16. CPAP.com offers this card: SD Memory Card for IntelliPAP Machines as an option. SD cards with other formats, such as an SDHC card formatted for FAT32, will not work with the IntelliPAP machine.
A bacteria filter will not install directly to the air outlet port on the IntelliPAP Heated Humidifier. The filter is too bulky and there is not enough space. IntelliPAP humidifier owners can order the FlexiTube Angle Adapter for CPAP and BiPAP Machines to provide a spacer between the humidifier and the bacteria filter. Your CPAP hose can then fit onto the other end of the bacteria filter.
5 Year Manufacturer IntelliPAP Machine Warranty
DeVilbiss offers an industry leading 5 (five) year manufacturer warranty on IntelliPAP machine orders placed on or after 2/15/2012.
3 Year Manufacturer IntelliPAP Humidifier Warranty
DeVilbiss offers a 3 (three) year manufacturer warranty on IntelliPAP Humidifier orders placed on or before 2/14/2012.
The following is a listing of IntelliPAP error codes and their meanings:
|E01||Read Setting Error. Stored prescription settings are corrupted. Unit is in fail-safe state (blower off) when this error displays. See Troubleshooting and Service Instructions to correct fault, or return unit for repair.|
|E03||Motor Error. Unit attempts to drive the motor, but the blower is not spinning (zero motor speed). See Troubleshooting and Service Instructions to correct fault, or return unit for repair.|
|E04||Locked Rotor Error. Average motor speed is low, while the motor current is high, for several seconds. See Troubleshooting and Service Instructions to correct fault, or return unit for repair.|
|E06||Motor Runaway Error. Motor speed is high and the pressure and total flow are low (close to zero) for a fixed period of time. See Troubleshooting and Service Instructions to correct fault, or return unit for repair.|
|E07||Motor Fault Error. The motor control chip fault output goes active low for a fixed period of time. See Troubleshooting and Service Instructions to correct fault, or return unit for repair.|
|EA0||Internal DataFlash Memory Failure. Internal DataFlash memory not recognized or not working. Unit must be returned for service. Critical error that will show up on the PAP LCD display. PAP Still functional.|
|EA1||Master Time Stamp Not Advancing. Real time clock not initialized or has failed. Unit must be returned for service. Critical error that will show up on the PAP LCD display. CPAP Still functional.|
Please note: The manufacturer warranty is voided by water damage or misuse of the machine.