300 Watt DC to AC Power Inverter allows any CPAP, BiPAP, VPAP or Bilevel device to run from a 12 volt cigarette lighter socket in a car, boat, or recreational vehicle.
Manufactured by Philips Respironics.
Product FeaturesThis device cannot be connected directly to the battery. You must use the Respironics, 12volt DC Battery Adapter Cable with Battery Clips (sold separately) and connect through the cigarette lighter connector on the battery clips.
- Powers AC products up to 300 watts
- High surge capacity for starting demanding loads
- Thermally regulated fan runs when needed
- Auto shutdown prevents total battery discharge
- Regulated output protects sensitive equipment
- One Year Warranty
- Max Continuous Load 300 watts
- Surge Capacity (peak) 500 watts
- Optimum Efficiency 90%
- No Load Current Draw 0.18 Amps
- Wave Form Modified Sine Wave
- Input Voltage Range 10-15 Volts DC
- Low Voltage Alarm Yes (10.7 Volts)
- Low Voltage Shutdown Yes (10.0 Volts)
- Overload Shutdown Yes
- AC Receptacle Dual
- Size (L x W x H) 6.0" x 4.7" x 1.8"
- Weight 1.8 Pounds
CPAP.com always offers these services:
- 8 am to 10 pm CST Toll Free Phone and Email Support
- 24 hour access to our CPAPtalk.com Support Forum
- Learning Center for information on sleep apnea and cpap equipment
- Consumer Purchasing Trends Reports
- Warranty Services for all items purchased through CPAP.com
- Insurance Compliant Invoices
- Free Prescription Request System
- Philips Respironics products are for sale only to patients within the U.S. and not to other third parties or businesses.
- All services provided by CPAP.com, including advertisement, ordering, sale, delivery, instruction, and setup of the products are performed solely by CPAP.com and independently of the manufacturer.
Important TipsThe 300 Watt DC to AC Power Inverter has an internal fan which uses a sensor to detect when it should turn on and off. The fan on this inverter will cut on and off as needed to maintain the desired temperature of the inverter.
The use of any inverter will reduce the battery life up to 50% because the inverter, itself, uses battery power. Other components like the humidifier will further reduce the run-time.
This inverter will not operate from a 6 volt battery.
CPAP Product Customer Reviews View All Reviews
No reviews to display
CPAP Video & FAQ
Frequently Asked Questions (FAQs)
- What type of power does my CPAP need?
- Most CPAP and BiLevel machines today are designed to operate on both AC and DC voltage. In the United States, the AC current from a wall outlet is 110 volts. Overseas, the AC current is usually 220 or 240 volt. Most batteries produce 12 volt DC current.
We include information about the power needs and capacities on the CPAP.com pages our products. Please see the "Power" tab on any machine product page for this information. You can also see a summary of the voltage range and whether it is DC capable on our Compare Charts.
Here is a video explaining differences in power cords and power supplies of various CPAP machines:
- Will my CPAP operate on DC / battery power?
- All CPAPs will operate on DC power with the use of an Inverter to change the current from DC to AC before it reaches the machine. However, many will operate on DC power without the need for an inverter, just a power cord that plugs into the CPAP and ends in a cigarette lighter plug.
Inverters do require a lot of power. If you need to use an inverter to run your machine on battery power, the inverter will probably take about half the life off the battery.
To determine if your CPAP will operate without an inverter, look for a DC outlet on the machine, either at the back or on the side, with a single opening directly in the center. It should be labeled "DC Power" or similar term. You can also review our Compare Charts to see if your machine has a optional DC cable option.
Here is a video describing different DC cable and battery options for various machines:
- Will my CPAP operate overseas?
- An adapter may be required to plug a US power cord into a wall outlet in a foreign country. CPAP.com sells an assortment pack of World Traveler Power Adapter Plugs, which allow CPAP machines to plug into different types of wall sockets found in different countries.
All the models CPAP.com sell contain power converters that enable the machine to operate on any AC current from 100 to 240 volts without any adjustment.
- How do I use a battery to operate my CPAP?
- All machines can connect to DC power. Some machines are configured to operate on DC Power and have a DC power outlet. Other machines require AC power and will need an Inverter that turns DC power to AC power.
Having the proper equipment is the biggest key to successfully operating the machine from a battery. For example, if your machine does have a DC outlet and you want to run your machine from a marine battery, be sure to get the correct DC cable for the machine, as well as an adapter cable with alligator clips that fits over the cigarette lighter plug.
CPAP.com sells complete battery kits with everything you need for many machines including the correct cables for those machines and batteries. You can see if there is a battery kit available for your machine Here. If you have questions about the specific power needs of your machine, review the 'Power' tab found on the machine's product page, or give us a call at 1-800-356-5221.
- What type of battery should I use?
- For longer battery use, we suggest using a deep-cycle marine battery to power the machine. A deep-cycle marine battery is heavier than the oridinary battery, but it has many advantages. The battery is easily found at auto shops and it provides power to the machine for an extended period of time. For example, a CPAP machine set at 10 cm H20 and charged on a deep-cycle marine battery will usually last three nights before it needs to be recharged.
Charging the machine from a car battery is another option. When charged from a car battery, the machine will last nearly eight hours before needing to be recharged. Please Note: It is possible that the car battery will not be able to start the next morning due to the draw of energy consumed by the machine.
There are small, portable battery packs that will operate a CPAP set at 10cm for at least one night, or about eight hours. CPAP.com sells full Battery Kits that come with everything needed to run your machine off battery power including the correct cables needed for your machine and a battery. You can find the battery kit for your machine Here. On each battery kit page, you can input your machine and pressure setting to see an estimate battery life for your machine.
- How do I prepare for power outages with CPAP?
- There are a lot of back up battery choices out there that can give you peace of mind in case of a power outage. The Transcend Travel CPAP Machine has Battery Options available that can set up as a fail-over, back up battery. Review the 'Power' Tab on the machine page.
CPAP.com sells Battery Packs to accompany your machine if an integrated battery is not available. Visit CPAPtalk for more advice specific to your needs, or view this past thread on the topic here.