Have a question? Call us!
Sat 8AM-5PM CST
Manufactured by Plastiflex Healthcare.
This is the replacement Hybernite Heated Tubing to be used with the Hybernite Rainout Control System. This product does not include the Power Supply Unit required to heat the tubing and is available: ="">Hybernite Rainout Control System.
The Hybernite Heated Breathing Tube is 71.25 inches from end to end including cuffs. Both cuff ends are ridged to avoid stress damage to the hose. The diameter of the hose is 19mm.Compatibility Notes:
- The Hybernite is not compatible with machine lines that use a smaller diameter (15mm) hose such as the S9 series by ResMed and the 60 Series by Philips Respironics.
- PR System One devices with software Versions EARLIER than 1.24 are not recommended for use with the Hybernite. The version of software flashes on the display screen when the machine is first turned on.
This Product Includes...
- Heated Tubing
- Length 71.25" (1.83m)
- Internal Diameter .3/4" (19mm)
CPAP.com always offers these services:
- 8 am to 8 pm CST Toll Free Phone and Email Support
- 24 hour access to our CPAPtalk.com Support Forum
- Learning Center for information on sleep apnea and cpap equipment
- Consumer Purchasing Trends Reports
- Warranty Services for all items purchased through CPAP.com
- Insurance Compliant Invoices
- Free Prescription Request System
Prescription: The Hybernite Rainout Control System and Hybernite Heated Hose require a prescription for purchase. If you don't have a prescription CPAP.com will gladly assist you in getting one. Visit our Prescription Policy for more information or call us at 1-800-356-5221.
A prescription that contains any of the following words is acceptable to make a purchase of the Hybernite: "Hose"; "Heated Hose"; "Supplies"; "Machine"; "CPAP"; "APAP"; "BiPAP"; "BiLevel"; "VPAP" or "Humidifier"
Compatible with ComfortLine Heated Tubing Kit
The Hybernite Heated Tubing is compatible with the ComfortLine Heated Tubing Kit. This tube can be connected to the ComfortLine power supply.
Power Supply Unit Location: The Power Supply Unit is not included with this product but can be purchase as the Hybernite Rainout Control System. The manufacturer recommends that the Power Supply Unit (PSU) for the Hybernite should be placed on the floor during operation.
Humidification Required: The Hybernite Heated Tubing must ALWAYS be used with water humidification to avoid elevated airflow temperatures.
Hose and Ambient Temperatures: The Hybernite hose will reach a temperature between 95°F (35°C) and 104°F (40°C) degrees without airflow from the machine. The Hybernite system is recommended to be used in a room with an ambient temperature between 41°F (5°C) and 86°F (30°C). The system should not be used in ambient room temperatures above 86°F (30°C).
Hose Covers: It is recommended that an insulated hose cover such as the SnuggleHose or other textile object NOT be used with the Hybernite Heated Tubing to avoid elevated temperatures.
Recommended Cleaning & Replacement Schedule: Prior to cleaning the power connection must be disconnected from the tubing. The tubing itself should be washed with mild soap and lukewarm water, then rinsed well, hung and allowed to dry. Tap the electrical connection end of the hose on a hard surface to release any water that could be caught inside. The tubing must be COMPLETELY dry before reconnecting to power. The manufacturer recommends replacing the Heated Tubing after 6 months of use.
Warranty: The Power Supply Unit carries a 1 year manufacturer warranty. The Tube carries a 30 day manufacturer warranty. CPAP.com will work with the manufacturer on your behalf. Please contact us at 1-800-356-5221 or by email at email@example.com.