IntelliPAP 2 AutoAdjust Auto CPAP Machine
IntelliPAP 2 AutoAdjust CPAP Machine Overview
DeVilbiss' IntelliPAP 2 AutoAdjust Auto CPAP Machine features comfort technology and OPTIONAL humidification while auto-adjusting pressure to meet various breathing events. The machine and OPTIONAL IntelliPAP 2 Standard Heated Humidifier offer a smaller footprint when stacked together.
The IntelliPAP 2 AutoAdjust Auto CPAP Machine from DeVilbiss is a travel-sized CPAP machine designed to automatically adjust pressure during the night. The machine uses multiple techniques to track therapy effectiveness including a SD card (included), bluetooth technology, and SmartCode.
- "PureView" AutoAdjust Algorithm
- Large LCD Screen Display
- SmartCode Adherence Score
- OPTIONAL Heated Humidifier
- Bluetooth Capable Technology
- SmartFlex & Adjustable Rounding
- Ramp Comfort Feature
- Auto Start / Auto Stop
"PureView" AutoAdjust Algorithm
The "PureView" AutoAdjust Algorithm is a unique feature used on the AutoAdjust CPAP machine to detect advanced apnea events such as: periodic breathing and flow limitation. This algorithm uses a logic that understands complex issues, leading to automatic changes in pressure in response to OSA and CSA events. This is an auto-adjusting machine that will operate between a low and high pressure range, remaining at the lower pressure until an event is detected requiring a higher pressure.
Large LCD Screen Display
The sharp LCD screen increases the visibility of displayed information and pairs with a knob that is turned and pressed for easy navigation of menus and settings. The display can also show regular reminders to change filters and clean equipment, if desired. When in standby, the machine will display a clock and also has an alarm clock function.
On-Screen Data Includes:
- Leak Rate
- Usage Information
SmartCode Adherence Score
SmartCode messages begin to display on the screen after the first night of therapy. Each code is a uniquely encrypted code holding personal sleep data results. SmartCode is intended to help IntelliPAP 2 users become compliant. Review the SmartCode Adherence Score with a physician.
SmartCode Adherence is determined after the machine is used for 30 consecutive days. Data can be generated to depict results from the following:
- 1 Day
- 7 Days
- 30 Days
- 90 Days
- A Cumulative Report
OPTIONAL Heated Humidifier
Devilbiss offers two integrated humidifiers for use with the IntelliPAP 2 machines. Warm, moist air travels through the hose to the mask to provide comfort to the user. The humidifiers attach to the base of the machine, so only the height is increased with the two are used together. The IntelliPAP 2 Auto machine can be paired with both versions of the IntelliPAP 2 Heated Humidifier:
- Standard Heated Humidifier: The standard humidifier can be set to a level ranging from 0-5 and will heat to provide that level of moisture all night long during inhalation and exhalation. Level 0 provides no heat, while level 5 provides the most heat.
- Pulse Dose Humidifier: The Pulse Dose feature provides moisture during inhalation only. During exhalation, the humidifier is bypassed, helping to clear moisture out of the hose, retain water in the chamber for a longer period of time, and reduce power draw from the machine and humidifier. The humidifier can be set from 0-5 with 0 being no heat and 5 being the highest heat setting.
Bluetooth Capable Technology
Enable the Bluetooth feature to begin pairing the IntelliPAP 2 to the SmartLink app designed by DeVilbiss. The SmartLink Data Management System is designed for use on Apple and Android devices. To begin using the app, visit the iTunes App Store or the Google Play Store depending on the device. Search "DeVilbiss SmartLink" to locate the SmartLink app. Once the app has been downloaded and the Bluetooth option is turned ON, data will begin streaming from the machine to the handheld device.
The SmartLink app is compatible with the following devices:
- iPhone, iPad, or iPod touch running Apple iOS 8 or newer
- Android smartphones and tablets running 4.0.3 or newer
SmartFlex & Adjustable Rounding
SmartFlex is designed to decrease pressure during exhalation 1, 2, or 3 cm for a more comfortable breathing pattern during therapy.
Adjustable rounding allows the user to customize the transition between exhalation to inhalation and inhalation to exhalation. The machine settings range from 0 to 5. Level 0 represents a direct jump in pressure, while level 5 represents the softest transition.
Ramp Comfort Feature
The IntelliPAP 2 AutoAdjust offers Ramp as a comfort feature. Ramp is designed to gradually increase therapy to the lower pressure of the auto range over a period of time. Between 0 to 45 minutes, the machine will slowly raise the pressure to help the user become acclimated to their therapy so they may fall to sleep with ease. The ramp button is on top of the machine and to the right of the humidifier symbol. Use the button to start or stop ramp. Please Note: If the machine’s lower pressure of the auto range is 3, ramp will not be available.
Auto Start / Auto Stop
The Auto Start / Auto Stop feature allows the user to begin and finish CPAP therapy without manually pressing a button. When enabled, place the mask against the face and begin breathing into the mask to start sleep therapy. Likewise, remove the mask when waking to stop sleep therapy. It may take a few moments for the "auto off" feature to kick in, and turn the device off. If the Auto Start / Auto Stop feature is not desired, the machine is equipped with button to manually control the device.
This Product Includes...
- IntelliPAP 2 AutoAdjust Auto CPAP Machine
- Carrying Case
- 6 Foot Long 15mm Slim Tubing
- Filter Cover
- Reusable Filter
- Disposable Filter
- SD Card
- Power Supply
- Power Cord
CPAP.com always offers these services
- 24 hour access to our CPAPtalk.com Support Forum
- Learning Center for information on sleep apnea and cpap equipment
- Consumer Purchasing Trends Reports
- Warranty Services for all items purchased through CPAP.com
- Insurance Compliant Invoices
- Free Prescription Request System
The user manual included with this machine can be viewed and downloaded here: IntelliPAP 2 Auto Machine.
Tips to Position the IntelliPAP 2 onto the OPTIONAL Humidifier
When placing the machine on top of the humidifier cradle, ensure the display is facing the same direction as the pale blue tab on the water chamber. Then, when removing the machine from the cradle, lightly push inward on the rectangular button near the rear corner of the cradle. When properly positioned, the units are easily attached and dismantled.
Compatible Replacement Parts
Please Note: Some of the parts used on the original IntelliPAP are also compatible with IntelliPAP 2 machines. Other parts are strictly designed for the IntelliPAP 2.
- These disposable filters and reusable filters are compatible with both the IntelliPAP and the IntelliPAP 2. However, the IntelliPAP 2 has a filter cover which was not used on the original machine.
- The IntelliPAP 2 comes with a 6 Foot Long 15mm slim hose. If the user prefers to use a longer hose or a hose with a 22mm diameter, the settings may be changed on the machine to accommodate the hose, with the exception of a heated hose.
- The IntelliPAP 2 has a power supply attached to the power cord. This design is different from the original power cord used with the IntelliPAP machines.
Air Supply Port Cover
The functionality of the device is dependent upon the placement of the air supply port cover. There are two ways to utilize the cover:
- When the IntelliPAP 2 machine and humidifier are connected, the air supply port cover folds into the circular crevice located on the back of the IntelliPAP 2 machine.
- If using the IntelliPAP 2 machine without the humidifier, reposition the air supply port cover so it coats the circular opening found on the bottom of the IntelliPAP 2 machine. Please Note: The cover must be put back into the opening found on the back of the machine when preparing to reconnect the machine and humidifier. If the cover is still protecting the hole on the bottom of the machine, the devices will not fasten.
Water Chamber Cleaning Tips
The water chamber for the IntelliPAP 2 Heated Humidifier needs to be cleaned daily. Please Note: The water chamber may be placed in the dishwasher for an occasional cleaning. When placed in the dishwasher, the chamber must sit on the top rack. For all other daily needs, hand wash the chamber with warm water and mild soap.
Before cleaning the chamber, the user will need to remove the tank from the humidifier cradle. Follow these steps to withdraw the chamber:
- Gently press on the chamber release latch which is located at the front of the chamber.
- Slide the chamber out of the cradle.
- Locate and push forward on the chamber disassembly lever which is directly behind the chamber release latch.
- The chamber will open into two halves.
Cleaning Tips for the Humidifier Cradle
The cradle which holds the water chamber should be cleaned weekly. Please Note: The cradle should not be submerged in water. Instead, use a moist cloth to wipe the exterior of the unit. Follow these steps when preparing the clean the cradle:
- Remove the power source to allow the heater plate to cool. The cooling process may take up to 10 minutes.
- Once cooled, withdraw the water chamber from the inside of the cradle.
- Use the flow generator release button to separate the IntelliPAP 2 machine from the cradle base.
- Use a moist towel to cleanse the cradle.
Instructions for Filling the Water Chamber
When refilling the water chamber, use distilled water to help prevent mineral build-up. The IntelliPAP 2 water chamber has water level indicators located on the sides and top of the chamber. Do not surpass the specified markings when refilling the chamber, as it may cause water damage.
Sealing Gasket Maintenance
Before each use, examine the sealing gasket surrounding the circumference of the chamber. Ensure the gasket has been put back into place after each cleaning.
Introducing the IntelliPAP 2
This video provides a demonstration and an introduction to the IntelliPAP 2 machine, going over basic features, as well as providing information on how to navigate through the menus and screens.
How to Connect the IntelliPAP 2 to Bluetooth on an Android Device
- Download the DeVilbiss Smart Link App for Your Android Device from the Google Play Store
- Open the App.
- Log Into the App Using Your Device Serial Number and Your Last Name. Don't forget to Capitalize!
- Turn the wheel to Select Add-Ons from the Menu
- Select Bluetooth from the List of Options
- Select "On"
- Leave the IntelliPAP screen on Bluetooth During Connection
- On Your Android Device, Tap the Continue Button. This will copy the PIN to your Clipboard.
- Under the Android Settings Menu, Select Bluetooth
- Locate a Device Begining with the Letters DV6, and Select it. This Will Bring Up a Place to Enter Your PIN.
- Check the Box Below that Says "PIN Contains Letters or Symbols"
- Paste the PIN into the box
- Click the OK Button
- On the IntelliPAP 2, Locate Your Android Device and Select It
The DeVilbiss IntelliPAP 2 system operates automatically between 110v and 240v AC. When traveling either inside or outside the USA, there are no switches to switch or fuses to change. The machine will automatically sense what voltage is needed and set itself to run on that power.
The user will need to adapt the plug that goes into the wall to fit the style of outlet socket that is being used. This is simply done with a Power Adapter Plug Kit. This is the same type of plug adapter that would be used with a hair dryer or an electric shaver. World Traveler Power Adapter Plugs.
The DC Power Cord for IntelliPAP 2 CPAP Machines plugs into the back of the IntelliPAP 2 units, allowing the user to connect the CPAP to an external DC battery source. The IntelliPAP can be powered by plugging the 12V DC Power Cord into a cigarette lighter plug. This is ideal for traveling by car, boat, or recreational vehicle that is equipped with a cigarette lighter plug.
Important Note: If using the OPTIONAL heated humidifier, the IntelliPAP 2 machines cannot be operated from DC power. A modified sine wave inverter of at least 120V AC is necessary to power both the machine and humidifier from a DC power source.
These are the compatible inverters:
Battery Backup Option
Because the IntelliPAP 2 machines use a separate connection for AC and DC power, they can be set up in a battery backup configuration. The AC power cord is plugged AC power source such as a wall outlet, and the OPTIONAL DC Power Cord for IntelliPAP 2 CPAP Machines is plugged into a DC power source such as a battery. The IntelliPAP will use the AC power source when available, otherwise will convert to the DC power source, without any interruption in therapy. If the AC power source becomes available again, the machine will revert back. Please note that if using the heated humidifier, it will not function from DC power.
The Portable Outlet is a great way to power this CPAP machine, as well as many others. The Portable Outlet converts the battery's stored DC power into convenient outlet power, allowing you to power any CPAP machine or electrical device using the main power supply that came with the device. The Portable Outlet is so versatile it can power TVs, stereos, blenders... anything! It also includes convenient USB ports that allow you to charge your phone or tablet while charging your CPAP.
Weighing in at only 1.7 pounds the Freedom Battery is a terrific option for travel. Pair the DC input cord included with the battery and the DC Power Cord for IntelliPAP 2 CPAP Machines to connect to the Freedom battery. This lithium-ion battery meets UN Manual Test and Criteria requirements for travel.
PLEASE NOTE: It is also possible to connect the IntelliPAP 2 to the Freedom Battery using the 150 Watt Sine Wave Inverter (sold separately).
The auto adjust feature goes into a “wide open” mode about every other night. It takes the pressure up to 20cm and then I have to turn it off and then back on to reset. I’m getting an “e81 pulse dose failure”. more...
Good design! High quality! Low noise. Covered air inlet with filter inside.
I had to return this machine (and a replacement) after both machines would wake me up in the middle of the night with full pressure (19+). I assume I had apnea incidents and the pressure went up but then stayed up. I would stay awake and breathe carefully but the pressure simply did not fall. There were not leaks in the tubing or mask; I've been using CPAP over 10 years. The machine I then bought was the ResMed AirSense 10 AutoSense and it works perfectly, showing a 90th percentile pressure level of 11 over many nights. CPAP.com was fantastic in helping me diagnose the problem and replacing the first machine and then permitting me to return the second IntelliPAP machine. more...
2 people found this helpful.
The size and features are great. I like the simplistic hose connection and good rubber base. I don't use the humidifier, so I can't comment on that connection. Using the rotating button for the menu is easy and the menu is well though out. My only gripe with it is how bright the buttons are and how long they stay bright. Even at the lowest brightness setting, it's too bright in the middle of the night when I have to get up. I would recommend this to anyone. more...
This is the first APAP machine I have ever used so this review is based on 6 months of use by a newbie. The unit is pretty much plug n play. I attached the pulsed humidifier and slipped in the water cartridge. Attached my mask and slept soundly for the first night. I really like how easily the cartridge is to remove and refill. I travel internationally 4 times a year for extended periods. The machine is small and super quiet. My wife is happy cuz she can't hear it or me snoring anymore! The unit itself is very well built and battle tested on two overseas trips using 110v at home and 220v abroad. The display is easy to use and has all the info and settings at your fingertips. Only knock I have is the iphone app software is clunky interfacing with the machine via bluetooth. Doesn't always download the machine therapy info to the app. You have to remove and reinstall the app on your iphone every once in a while to get the app software synced up with the machine. Overall I am extremely happy with this purchase from cpap.com more...
Compare Similar Products
Basic FeaturesManufacturerDeVilbissResMedResMedTypeAPAPAPAPAPAPWarranty3 Years2 Years2 YearsSound Level<26.6 dBA26.6 dBA30 dBAQuietYesYesNoManuals IncludedUser ManualPatient/Quick SetupPatient
DimensionsTravel SizedYesYesYesUnder 3 lbsYesYesYesMachine Weight1.93 lbs2.750.66 lbsEntire Weight5.87 lbs5.1 lbs2.06 lbsEntire Size11.7 in L x 7.6 in W x 3.10.04" x 4.75" x 5.91"NAEntire Package Size11.75 in L x 8 in W x 10.13.24" x 10" x 5"12.25 in L x 4.6 in W x 3Machine Size Only6.1 in L x 5.9 in W x 3.7NA5.4 in L x 3.3 in W x 2.1
Humidifier FeaturesHeated HumidifierIntelliPAP 2 HeatedHumidAirNoBuilt-In HumidifierNoYesNoHeated Humidifier TypeIntegratedBuilt InNAAmbient Rainout ReductionPulse DoseAutoClimateControlNoIntegrated Heated HoseNoClimateLineAireNoHumidifier Water Capacity375 mL380mLNA
Pressure FeaturesAuto Pressure AdjustmentYesYesYesPressure Range3 cm H2O to 20 cm H2O4-20cm H2O4-20 cm H2OEPAP Pressure RangeN/ANANAEasy BreathingSmartFlexAutoSet with EPRAutoSet with EPRRampYesYesYesCPAP ModeYesYesYesLeak CompensationYesYesNoMax Operating Altitude9, 800 ft8,500 ft8,500 ft
Power FeaturesDirect Battery OperationYesYesNoVoltage Range100 - 240 V AC100-240V100-240V ACOptional DC CableYesYesYesIntegrated BatteryNoNoNo
Software FeaturesAdvanced Software/DataMSD; OSYesYesData CardYesYesNoOptional SoftwareSmartLinkResScanAirMini App
Additional FeaturesAuto Altitude AdjustmentYesYesYesAuto ON/OFFYesYesYesMask Off AlertLCDNoNoSpontaneousNoNoNoTimedNoNoNoDesigned For HerNoNoNoBack Lit DisplayYesYesNoBacklight Shuts OffYesYesNA