PR 60 Series BiPAP autoSV Advanced Overview
The PR System One REMstar 60 Series BiPAP Auto SV Advanced has all the best features of the PR System One REMstar BiPAP Auto SV Advanced along with the option of an integrated heated hose. It combines the support of a BiPAP machine and the accuracy of Auto EPAP technology to monitor the individual and continuously change based on needs. Cases of Complex Sleep Apnea can be managed by the system constantly adapting to individual inhalation and exhalation needs and using the titration algorithm to appropriately ventilate every breath.
The PR System One REMStar 60 Series BiPAP autoSV Advanced is a designed to manage complex sleep-disordered breathing with sophisticated technologies.
- Auto EPAP
- Advanced Back Up Rate
- Servo Ventilation (SV) Algorithm
- BiFlex Comfort
- System One Resistance Control
- Optional 60 Series Heated Tube Humidifier
Auto EPAP technology distinguishes between obstructed and clear airway apneas. Using an Auto-titrating algorithm allows the device to operate like an Auto CPAP or Auto BiPAP device and adjust when needed to maintain the airway. Auto EPAP adapts to changes in the user's needs such as a change in weight or medication.
Advanced back-up rate
Complex Sleep Apnea occurs when Central Sleep Apnea is experienced while being treated with a CPAP or BiPAP Machine for Obstructive Sleep Apnea. The Advanced Back Up Rate of the autoSV helps to prevent this by encouraging spontaneous breathing at a natural rate for the user. This is accomplished by the system automatically calculating the backup rate to closely match the natural breathing rate.
Servo ventilation (SV) algorithm
The SV algorithm provides a smoother breathing pattern. By monitoring peak flow and changing pressure as needed on a breath by breath basis it supports a proper, stable breathing cycle.
Bi-Flex technology provides constant pressure relief to enhance the comfort of therapy and provide a more natural feel. Pressure is reduced by half centimeter increments during the transition from exhalation to inhalation, inhalation to exhalation and during exhalation.
System One Resistance Control
System One Resistance Control considers the resistance of the mask itself to help attain optimal pressure. PR employs a numbering system on all of its masks to reflect the variables of the mask and the number is entered at the time of set up. This compensates for the variance in pressures in each mask and provides a more consistent pressure.
Optional System One Humidity Control and Integrated Heated Tube
The PR System One REMStar 60 Series BiPAP can be used with the optional Heated Tube and System One Humidity System to maximize comfort, manage rainout and reduce the potential of water damage. The machine adjusts the relative humidity delivered to the mask by analyzing the ambient room temperature and machine airflow. The optional slim 15mm Heated Tube maintains the desired temperature evenly from the machine to the mask.
When the optional 60 Series Heated Humidifier and Heated Tube are added to the machine the on-screen menu offers a selection of three available levels of relative humidity (70, 80 or 90 percent). Adjust the control dial to choose from 5 levels of heated tube temperature, ranging from 4 to 16 degrees above room temperature. The optimum humidification level at the mask is easy to maintain when using the PR System One REMStar 60 Series humidification assembly.
The chamber used with the System One 60 Series Humidifier has increase in surface area and greater level of humidification output. The Heated Tube easily snaps on and off the humidifier port.
The potential for water damage is greatly reduce by System One Dry Box technology. Unique design features prevent water from exiting the chamber and entering the machine. The center air outlet port uses a one way valve. This location on the chamber helps to minimize total water displacement. If water were to escape the chamber it is trapped within the reservoir of the Dry Box.
The PR System One 60 Series BiPAP autoSV comes standard with carrying case, memory card, manuals and Respironics Performance Tubing. If the PR System One 60 Series Heated Tube Humidifier with Heated Tube is purchased it will be package the heated tube.
This Product Includes...
- 60 Series BiPAP autoSV Advanced Machine
- Carry Case
- System One REMStar 60 Series Power Supply and Cord
- SD Memory Card
- Reusable Foam Filter (2)
- Patient Manual
- Clinical Quick Setup Guide
Pressure Range: IPAP - 4-25 cmwp; EPAP - 4-25 cmwp (.5 increments)
Ramp Time: 0 to 45 min. (5-min. increments)
Starting Ramp Pressure: 4 to EPAP; patient-adjustable
Dimensions - Machine Alone: 7 in L x 5.5 in W x 4 in H (18 cm L x 14 cm W x 10 cm H)
Dimensions - Machine with Heated Tube Humidifier: 10.75 in L x 6.5 in W x 4 in H (27 cm L x 16.5 cm W x 10 cm H)
Weight - Machine Alone: 3 lbs / 1.36kg
Dimensions of 60 Watt Power Supply (For Use Without Heated Tube) - Not Including Cord or Plug: 4.75 in x 2.75 in x 1.75 in
Weight of 60 Watt Power Supply - Not Including Cord: 14.72 oz (0.92 lbs)
Dimensions of 80 Watt Power Supply (For Use With Heated Tube) - Not Including Cord or Plug: 6 in x 2.75 in x 1.75 in
Weight of 80 Watt Power Supply (For Use With Heated Tube) - Not Including Cord: 1 lb 3.04 oz
AC Power Consumption (with 60 W power supply): 100 - 240 VAC, 50/60 Hz, 2.1 A max.
AC Power Consumption (with 80 W power supply): 100 - 240 VAC, 50/60 Hz, 2.0 A max.
DC Power Consumption: 12 VDC, 6.67 A
Altitude Compensation: Automatic - Maximum height 7500 feet.
Filters: Foam and optional ultra-fine
Device Set-Up: LCD/control wheel/push button
Data Storage Capacity
Compliance Meter: Breathing detection
Altitude compensation: Automatic
Electrical Requirements: 100 - 240 VAC, 50/60 Hz, 12volt DC
European Voltage Usage: Yes, outlet adapters required
Warranty: Two years; replace with new
Max Exposure Heat for Machine: Machine should not be exposed to heat greater than 140 degrees Fahrenheit
Sound Level - Mask Leak Alert: 35 dB, +/- 2 dB
Required Settings: This machine does not have required settings.
Unless otherwise directed, CPAP.com will leave the PR System One REMStar 60 Series BiPAP autoSV Advanced at the default settings of:
- Max pressure: 25 cm H2O
- EPAP Min: 4 cm H2O
- EPAP Max: 15 cm H2O
- BPM: Auto
- Bi-Flex: 2
- Auto-On: On
CPAP.com always offers these services
- 24 hour access to our CPAPtalk.com Support Forum
- Learning Center for information on sleep apnea and cpap equipment
- Consumer Purchasing Trends Reports
- Warranty Services for all items purchased through CPAP.com
- Insurance Compliant Invoices
- Free Prescription Request System
- Philips Respironics products are for sale only to patients within the U.S. and not to other third parties or businesses.
- All services provided by CPAP.com, including advertisement, ordering, sale, delivery, instruction, and setup of the products are performed solely by CPAP.com and independently of the manufacturer.
- A valid prescription is required for the purchase of this product.
Important Compatibility Notes:
The 60 Series Heated Humidifier that interfaces with PR System One REMStar 60 Series Machines CANNOT be used with any other machines made by Philips Respironics including other versions of the PR System One REMStar line of machines. It is compatible only with System One REMStar 60 Series Machines.
There are physical characteristics which identify the PR System One REMStar 60 Series Machines and Heated Humidifiers. The humidifier used with the heated tube cannot physically be connected to any other versions of machine in the PR System One line. The heated tube is 72 inches in length.
Identifying PR System One REMStar 60 Series Machines
- PR System One REMStar 60 Series Machines and Heated Humidifiers are charcoal gray with metallic silver side and instrument panels.
- Other PR System One units are dark gray with ivory colored side and instrument panels.
- The PR System One REMStar 60 Series Machines have the words "System One" in raised or embossed letters stamped in silver on both the machine and humidifier.
- Other PR System One units do not have raised lettering, the branding information is painted onto the unit.
- Machine and Humidifier Connections
- PR System One REMStar 60 Series Machines and Heated Humidifiers have a 6 pin connection between the machine and humidifier.
- Other PR System One Units have a 4 pin connection between the machine and humidifier
- PR System One REMStar 60 Series Machines and Heated Humidifiers have a mechanical connection between the machine and humidifier
- Other PR System One Units have 2 tabs on the humidifier that fit into 2 slots on the machine.
- Model Number Reference
- Label identification example: the label on the bottom of the PR System One REMStar 60 Series Auto CPAP lists: "REMStar Auto"; "REF" of 560P; and Serial Number identified by "SN". The number 560 references the part number of the machine.
- The label on the bottom of other versions of System One machines lists: Machine name; "REF" that does not have the number "6" in the second position from the left; and Serial Number identified by "SN". For example the label on the bottom of a PR System One REMstar Auto CPAP Machine with A-Flex lists "REMStar Auto" and "REF" of 550P. The number 550 references the part number of the machine.
The PR System One Heated Tube for 60 Series Machines is a heated hose that integrates with the 60 Series Heated Humidifier. The heated tube connects to electrical ports on the humidifier.
Not using the Heated Humidifier? The heated tube can still be used by connecting to ports on the air outlet on the side of the machine. In this configuration, the hose will not heat and will act as a standard hose.
In addition to this heated tube that works with the System One 60 Series Machines, a traditional CPAP hose can be used. Set the size of hose you will be using in the Set Up Menu under "Tubing Type". Select either 15 mm or 22 mm depending on the size tube being used.
System One Resistance Control Settings
Please reference this link to find the resistance control setting for the different Philips Respironics masks: System One Resistance Control Settings.
Side Cover Panel
PLEASE NOTE: The manufacturer does NOT include the side cover panel when a 60 Series machine and Heated Humidifier are purchased together as a set. This part may be purchased separately: Side Panel for PR System One 60 Series CPAP Machines. If the machine is used without the humidifier, this plate may be snapped onto the side of the machine.
Optional System One 60 Series Heated Tube Set-Up
Philips Respironics provides these instructions for setting up the heated tube: System One 60 Series Heated Tube Set-Up
PR System One Humidifier Modes PR System One REMstar 60 machines version 1.22 or higher has two humidification modes:
- Classic Humidification
- System One Humidification
Version Note: To determine the version of software installed on a PR System One REMStar 60 machine, unplug the machine and plug it back into an outlet. The version is displayed on the screen when the unit is first plugged into an outlet. The humidification settings are displayed only when a PR System One Heated Humidifier is attached to the machine.
Classic Mode Humidification
The Classic mode of humidification allows the CPAP user to control the heat setting on the humidifier. The user will select "Classic" mode settings C1,C2, C3, C4 or C5. The "C" in front of the humidification setting indicates "Classic". C1 is the lowest level of humidification and C5 is the highest level of humidification.
Please Note: If the PR System One Heated Tube for 60 Series Machines is attached to the humidifier the Classic mode humidification will not be offered, only System One Humidification mode. To use the Classic mode of humidification attach a non-heated tube to the humidifier.
System One Humidification Mode
The System One Humidification uses the technology of the system to set the level of humidification delivered. The machine makes the decision based on the humidity present in the room air. The CPAP user decides how much they want to receive. A setting of 1 delivers the lowest humidity level and a setting of 5 delivers the highest level. The humidification setting will show 1 - 5. The letter "C" will NOT precede the number.
Passover Humidification, Heat Off
To turn the heated humidifier off, set the humidifier to zero "0". The air from the machine will pass over the water in the chamber acting as a passover Humidifier.
Setting the Humidification Mode and Level:
- The humidifier menu will only show when the humidifier is attached to the machine.
- Using the wheel on the machine, select "Setup" menu.
- Scroll down the menu offerings to locate: "SYSTEM ONE humidification".
- Set to "On" to allow the technology of the machine to set the amount of humidification delivered to the system based on the setting of 1 -5. When in the SYSTEM ONE humidification mode there is no letter indicator next to the number.
- Set to "Off" to control the heat setting manually. The letter "C" is displayed in front of the humidification number to indicate "Classic". C1 is the lowest heat setting and C5 is the highest heat level.
- Set the humidification level by adjusting the number to the right of "Humidifier", the menu item below SYSTEM ONE humidification. 1 is the least and 5 is the most humidification.
Power Supply and DC Cable
Please Note: If the 60 Series BiPAP autoSV Advanced machine is purchased individually it will include a 60 Watt power supply. If the 60 Series BiPAP autoSV Advanced machine is purchased with the heated tube humidifier and heated tube, it will include an 80 Watt power supply. The 60 Watt power supply will not power the heated tube. If using the Heated Tube Humidifier and Heated Tube, the 80 Watt power supply must be used to successfully heat the tube. The 60 watt supply weighs 10.95 oz and is 4.5" L x 2.75" W x 3" H with the cord wrapped around. The 80 watt supply weighs 1 lb 3 oz and is 6" L x 2.75" W x 3.25" H with the cord wrapped around.
The External 80 Watt Power Supply for PR System One REMstar 60 Series Machines is NOT compatible with other versions of PR System One machines. The connection peg that plugs into the PR System One REMStar 60 Series Machine is a specific size and will fit only this machine. The external power supply for this machine has different specifications than the External Power Supply for PR System One REMstar CPAP and BiPAP Machines used with other machines in the Philips Respironics lines. The Power Cord that plugs the external power supply into the wall outlet is the same for all System One Machine.
The Shielded DC Cord for PR System One 60 Series Machines that is compatible with PR System One 60 Series Machines works ONLY PR System One 60 Series Machines. The 12 Volt DC Power Cord (connects CPAP to cigarette lighter socket) used with other PR System One machines IS NOT compatible with PR System One 60 Series Machines.
This machine comes with a 2 Year Manufacturer Warranty. CPAP.com will work with the manufacturer on your behalf if the machine needs repairs. Please follow these steps:
- Contact us at email@example.com or 1-800-356-5221 to receive an RMA (Return Merchandise Authorization) Number before returning the equipment
- Include a note with the machine describing the issue you are experiencing, along with your full contact information to include a return shipping address and the RMA number provided to you
- Ship only the defective items to CPAP.com at 13235 N Promenade Blvd Stafford, TX 77477
- Do not return accessory items such as the carry bag, hose or power cords
- Email updates will be sent to keep you informed as the machine moves through the warranty process
- CPAP.com will contact the manufacturer on your behalf and handle all shipping between us and the manufacturer
- The repaired or replaced unit will be returned to you. Shipping charges may apply.
Please note: The manufacturer warranty is voided by water damage or misuse of the machine.
Humidifier Modes PR System One Heated Humidifiers Version 1.22 or higher have two modes:
- System One Humidification
NOTE: The version of software installed on a machine is displayed on the screen when the unit is plugged into an outlet. The Humidification settings are displayed only when a PR System One Heated Humidifier is attached to the machine.
The CPAP user who wants to control the heat setting on the humidifier should select "Classic" mode settings C1,C2, C3, C4 or C5. The "C" in front of the humidification setting indicates "Classic". This allows you to set the level of humidification.
System One Humidification Mode
The CPAP User who wants to allow the technology of the System One Humidification to set the level of humidification delivered based on the humidity present in the room air should select "System One Humidification". The humidification setting will show 1 - 5 but WITHOUT a "C" preceding the number.
If set to zero "0", the humidifier will act as a Passover Humidifier.
Accessing Humidification Settings:
- Humidifier must be attached to the machine
- Access the Set Up menu
- Scroll down the menu offerings to locate: "System One Humidification"
- If set to "On", the technology will set the amount of humidification delivered to the system based on the setting of 1 -5 with 1 being the least and 5 being the most
- If set to "Off", the user controls the heat setting based on C1-C5 with C1 being the lowest heat and C5 being the highest heat.
This system operates automatically between 110v and 240v AC. When traveling either inside or outside the USA, there are no switches to switch or fuses to change. The machine will automatically sense what voltage is needed and set itself to run on that power.
The user will need to adapt the plug that goes into the wall to fit the style of outlet socket that is being used. This is simply done with a Power Adapter Plug kit. This is the same type of plug adapter that would be used with a hair dryer or an electric shaver. World Traveler Power Adapter Plugs.
Optional DC Cable
The machine can operate using DC Power using the Shielded DC Cord for PR System One 60 Series Machines. This cord is compatible ONLY with PR System One 60 Series Machines. The 12 Volt DC Power Cord (connects CPAP to cigarette lighter socket) used with other PR System One machines IS NOT compatible with PR System One 60 Series Machines.
The Freedom Travel CPAP Battery is currently the best battery on the market to power your 60 Series machine. It makes a native connection without needing to buy an inverter or extra equipment to make it work. The Freedom is the next generation of the C-100 travel battery and is designed to work well with your 60 Series.
This portable battery pack is an external lead acid battery that is the ultimate in power, portability and convenience. This battery kit comes with the Respironics battery and the Shielded DC Cord for PR System One 60 Series Machines.
Compare Similar Products
Basic FeaturesManufacturerRespironicsPhilips RespironicsResMedTypeBiPAP AutoSVBiPAPBiPAP AutoWarranty2 Years2 Years2 YearsSound Level29 dB25.8 dBA26.6 dBAQuietYesYesYesManuals IncludedPatientPatientPatient/Quick Setup
DimensionsTravel SizedYesYesYesUnder 3 lbsYesYesYesMachine Weight2.8 lbs1.56 lbs2.75 lbsEntire Weight6 lbs4.37 lbs5.1 lbsEntire Size6.5" x 10.75" x 4"11.7" x 7.6" x 3.3"10.04" x 4.75" x 5.91"Entire Package Size8 " x 14.75" x 8"15" x 8.5" x 5.5"13.24" x 10" x 5"Machine Size Only7" x 5.5" x 4"6.2" x 7.6" x 3.3"NA
Humidifier FeaturesHeated HumidifierSystem One 60 SeriesDreamStation HumidifierHumidAirBuilt-In HumidifierNoNoYesHeated Humidifier TypeIntegratedIntegratedBuilt-InAmbient Rainout Reduction60 Series Heated TubeDreamStation Heated TubeAuto ClimateControlIntegrated Heated HoseYesYesClimateLineAirHumidifier Water Capacity325 mL325 mL380 mL
Pressure FeaturesAuto Pressure AdjustmentYesYesYesPressure Range4-25cm H2O4 - 25 cm H2O4-25 cm H2OEPAP Pressure Range4-25cm H2O4 - 25 cm H2O3-25 cm H2OEasy BreathingBi-FlexBi-FlexEPR with Easy-BreatheRampYesYesYesCPAP ModeIPAP=EPAPYesYesLeak CompensationYesYesYesMax Operating Altitude7,500 ft7,500 ft8,500 ft
Power FeaturesDirect Battery OperationYesYesYesVoltage Range100-240V AC100 - 240 V AC100-240VOptional DC CableYesYesYesIntegrated BatteryNoNoNo
Software FeaturesAdvanced Software/DataMSDMSD, OSYesData CardYesYesYesOptional SoftwareEncoreViewerDreamMapperResScan
Additional FeaturesAuto Altitude AdjustmentAutoYesYesAuto ON/OFFYesYesYesMask Off AlertLCD; ALARMNoNoSpontaneousNoNoYesTimedNoNoNoDesigned For HerNoNoNoBack Lit DisplayYesYesYesBacklight Shuts OffYesYesYes