Item # DS6T
Not Currently In Stock
Item # DS6T
Not Currently In Stock
The Respironics System One Humidifier with Heated Tube optimizes air stream humidification and reduces rainout, helping you get better sleep night after night.
Respironics System One Humidifier with Heated Tube Features and :
Enhance Your Sleep Therapy with Three Levels of Humidification for Optimum Humidity
Offering three levels of humidification-70%, 80%, and 90%-the device has five temperature levels ranging from 4 to 16 degrees above the ambient temperature to ensure optimum humidification for your CPAP therapy.
Outfitted With Dry Box Technology and Valves to Prevent Trapped Water
Included with the Respironics System One Humidifier is Dry Box Technology, which traps water in the humidifier and prevents it from entering the machine. The one-way valve on the outlet port also helps prevent water from exiting the humidifier water chamber, as well.
Works With Certain CPAP Machines from Philips Respironics Only
The PR System One 60 Series Heated Tube Humidifier is compatible with the following machines from Philips Respironics only:
This device is not compatible with other machines manufactured by Philips Respironics, including the System One REMstar Machines, although it can work with the 60 Series SE machine at reduced capability-it only delivers 80% humidity-and the humidity level is not adjustable.
Atmospheric Pressure: 77 to 101 kPa (0 - 7500 ft.)
Standards Compliance: This humidifier is designed to conform to the following standards: IEC 60601-1 General Requirements for Safety of Medical Electrical Equipment EN ISO 8185 General Requirements for Humidification Systems
Type of Protection Against Electric Shock: Class II Equipment
Degree of Protection Against Electric Shock: Type BF Applied Part
Equipment Type: IPX1: Drip Proof Equipment
Mode of Operation: Continuous
Electromagnetic Compatibility: The device meets the requirements of EN 60601-1-2, 2nd edition.
Pressure Drop with Humidifier: (max.) 0.3cm H2O at 60 LPM flow
Humidity min Output: 10 mg H2O/L, measured at max flow, 35º C, 15% RH
Disposal: When necessary, dispose of the humidifier in accordance with local regulations.
Heated Tubing Specifications
Maximum Recommended Pressure: 25 cm H20
Power Supply and DC Cable
The External 80 Watt Power Supply for PR System One REMstar 60 Series Machines included in this product is NOT compatible with other versions of PR System One machines. The connection peg that plugs into a PR System One REMStar 60 Series machine is a specific size and will fit only these machines. The external power supply for these machines has different specifications than the External Power Supply for PR System One REMstar CPAP and BiPAP Machines used with other machines in the Philips Respironics lines. The Power Cord that plugs the external power supply into the wall outlet is the same for all System One machines.
Please Note This product includes:
Philips Respironics packages their System One 60 Series machines sold without a humidifier with a 60 Watt External Power Supply. The 60 Watt power supply will not power the heated tube. When using the heated tube the 80 Watt External Power Supply included in with this humidifier must be used to successfully heat the tube. If a heated tube is not heating, check to be certain the 80 Watt power supply is being used.
Philips Respironics is developing a DC power cord for the 60 Series Machines but it is not yet available. When the DC power cord compatible with the PR System One 60 Series Machines is available it will be carried by CPAP.com. The 12 Volt DC Power Cord (connects CPAP to cigarette lighter socket) used with other PR System One machines is NOT compatible with the PR System One 60 Series Machines.
Identifying PR System One REMStar 60 Series Machines
There are physical characteristics that identify a PR System One REMStar 60 Series Machine. The humidifier capable of being used with the heated tube cannot physically be connected to any other version of machine in the PR System One line. The heated tube is 72 inches in length.
System One 60 Series Heated Tube Set-Up
Philips Respironics provides these instructions for setting up the heated tube: System One 60 Series Heated Tube Set-Up
Please note: The manufacturer warranty is voided by water damage or misuse of the machine.
In addition to the Heated Hose that works with the System One 60 Series Machines, a traditional CPAP hose can be used. Set the size of hose you will be using in the Set-Up Menu under "Tubing Type". Select either 15 mm or 22 mm depending on the size tube being used.
Maximum Fill Level Tips
Philips Respironics System One Humidifier Modes PR System One REMstar 60 machines version 1.22 or higher has two humidification modes:
Version Note: To determine the version of software installed on a PR System One REMStar 60 machine, unplug the machine and plug it back into an outlet. The version is displayed on the screen when the unit is first plugged into an outlet. The humidification settings are displayed only when a PR System One Heated Humidifier is attached to the machine.
Classic Mode Humidification
The Classic mode of humidification allows the CPAP user to control the heat setting on the humidifier. The user will select "Classic" mode settings C1,C2, C3, C4 or C5. The "C" in front of the humidification setting indicates "Classic". C1 is the lowest level of humidification and C5 is the highest level of humidification.
Please Note: If the PR System One Heated Tube for 60 Series Machines is attached to the humidifier the Classic mode humidification will not be offered, only System One Humidification mode. To use the Classic mode of humidification attach a non-heated tube to the humidifier.
System One Humidification Mode
The System One Humidification uses the technology of the system to set the level of humidification delivered. The machine makes the decision based on the humidity present in the room air. The CPAP user decides how much they want to receive. A setting of 1 delivers the lowest humidity level and a setting of 5 delivers the highest level. The humidification setting will show 1 - 5. The letter "C" will NOT precede the number.
Passover Humidification, Heat Off
To turn the heated humidifier off, set the humidifier to zero "0". The air from the machine will pass over the water in the chamber acting as a passover Humidifier.
Setting the Humidification Mode and Level:
Yes. Federal law requires we have a valid prescription on file before we ship your mask, machine, humidifier, or EPAP therapy, including Provent and Bongo RX. You will also need a prescription to purchase the Nightbalance POSA device.
We provide many easy ways to get your prescription:
For more information on prescriptions, please see our Prescription Section of the Learning Center.
Overall, this replacement works well. Only complaints: heated tube comes apart from my mask more easily than previous tube. Also, unless I haven't looked close enough, the water container doesn't have a fill line.
I love my 60 series CPAP, but wish i could get the replacement heated tube. i guess I'll have to shop around.
Vibration and leakage at the hose fitting within three years. I glued it fixed and put a rubber band around it all, but the machine-humidifier combination got louder and louder. Time for a new machine, but the humidifier failed first. BTW I do not use the heated hose feature but that does not matter..
My long-used humidifier stopped working. I ordered this replacement. Then I realized I had inadvertently changed the humidifier settings on the CPAP machine. Reset those, and the old one worked fine. I sent the new one back in the unopened box. The real story here is how the company treated me--with kindness and helpfulness--refunded my purchase price minus a 15% restocking fee (fair enough) and my paying the return postage. This has been a great company to work with for the many years that I have been a customer.
I bought this as a replacement for the original humidifier and heated hose, because the hose was not heating. The replacment hose is barely heating anymore than the one I am replaceing. During the first couple of winter seasons, the heating of the first hose was pretty noticeable, with the settings for the hose at the highest level. Now with this replacement one with the settings at the highest level, its not the same as before.
Life is so much better now..
It was a pleasure to have this product, it can with everything I needed to use it but unfortunately it was no useful for me, because it did not revolve my problem. My problem is that my CPAP had not been warm the water up. I will need a new CPAP.
Another super purchase to go with my new CPAP machine. Love CPAP.com for their excellent prices and speedy service.
The heated tube came with the humidifier. It makes the incoming air too warm for me when I increase the humidity, as tube temp goes up at the same time. Preferred my old non-heated tube. The humidifier works fine.
This humidifier works great and does what my previous two have done - humidify my high desert CPAP air. I just notice that each model has gotten cheaper. With the first, the water tank was all clear plastic and marked in black for easy inspection. The next model used a cheaper gray plastic for the lid. Now this model has the gray lid and they have removed the black line marking the maximum water level. Cost shaving and profit increase at it best.
Excellent machine and good value.
1 - 12 of 341 Reviews
Show 8 More