The backlight for this machine either shuts off automatically after a short intreval or can be set to off.
The machine display is lit for ease of reading.
These machines were designed with a woman's lifestyle in mind.
Timed (T) or Spontaneous Timed (ST) modes provide a customizable backup rate that forces the machine to switch from exhalation to inhalation. In ST Mode, the machine monitors the breathing cycle and switches from EPAP to IPAP once the backup time is met or whenever the break is detected through the sensors; whichever occurs first. The time aspect is only meant to act as a backup respiratory rate to the (S) function (see Spontaneous). This is used for patients with central sleep apnea, or those who need non-invasive mechanical ventilation.
Spontaneous (S) mode tells the machine to start blowing at IPAP pressure with no specific interval set. The machine must wait and detect a break in the breathing cycle before reacting, switching from exhalation to inhalation. If a backup rate is needed, then the unit will need a time interval setting and will be marked 'Yes' on the Timed property.
This feature detects excessive mask leaks due to poor cushion seal or when a user stops therapy by removing a mask when in use. The alert can be in the form of an audible alarm (ALARM) or a text message on the unit's LCD display (LCD).
This feature turns the machines off and on when putting on or taking off the mask. When you put your mask on the machines senses you breathing and turns itself on. Take off the mask and the machine turns off.
Altitude Adjustment is defined as the machine's ability to compensate for changes in altitude. Compensation can either occur manually or automatically. Manual adjustments occur within the Setup Menu and must be set by the user. Automatic requires no user intervention.
Software is an option on some CPAP machines. The software can give details on compliance and performance. The patient or physician downloads the data from the CPAP machine to determine how long a patient has used their machine each night and how well the machine is working to stop Apnea/Hypopnea events. This property shows the software package to use for machine that has the ability to transfer data to your computer and report Hours of Usage. See Advanced Software for a list of machines that can report AHI or Leakage.
This property marks the machines that have the ability to record performance data on a removable card. Some machines that are flagged as having software (seeOptional Software), do not have removable memory cards, rather instead, records information on-board and must use a download cable to retrieve the data from the machine.
Identifies which machines are data capable and have the ability to track your therapy progress over time. The machine captures key metrics like AHI (Apnea/Hypopnea Index), Leakage, and most pressure used. Depending on the machine, the data can be retrieved through an on Machine Screen Display (MSD), through Optional Software (OS), and/or through SmartCode (SC).
An integrated battery pack attaches to and functions seamlessly with the machine it is integrated with. It allows your CPAP to run on AC and fail over to the DC battery only if necessary.
A 'Yes' answer indicates that the CPAP Machine has an optional DC cable that plugs directly into a DC port. This allows you to use a DC power source, such as a battery or car cigarette lighter outlet. Some optional DC cables include alligator clips to connect directly to battery terminals. See the "Power" tab of the machine's product page for more detailed information on how the machine works with alternate power sources, and what optional cables and batteries are compatible.
Domestic Voltage is the voltage that is used with a standard wall outlet in the United States and ranges from 100-120 volts. International Voltage can range from 220v-240v AC.
Indicates which machines have a DC port in which you can use to connect a battery pack or a deep cycle marine battery with an optional DC cable. This is different from an integrated battery that works only with the machine it was designed for. (i.e. Everest CPAP Machine) see property Integrated Battery
This is the maximum altitude at which the machine will properly operate.
Leak Compensation is a machine feature that identifies mask leaks and adjusts to maintain the delivered pressure as set. If marked N/A assume the information is not available due to being out of production or unavailable.
This property defines whether the machine is capable of continuously blowing a constant pressure. All CPAP machines are defined as 'continuous', but some BiPAP's and APAP's also come with a CPAP mode; some BiPAP and BiLevel machines Do NOT. By setting the IPAP and EPAP to the same pressure, a CPAP Mode is achieved, preventing the EPAP pressure from dropping.
The Ramp feature allows the user to start treatment at a lower pressure to fall asleep with, gradually raising the pressure over a set time. This is a comfort setting and can set from 0 to 45 minutes on most CPAP machines.
Pressure Relief is a feature that lowers your therapy pressure to help with difficulties breathing against the machine's airflow, providing a more comfortable and natural breathing pattern. Assistance can occur on the Exhalation, on the Inhalation, and on both exhalation and inhalation at the same time.
This Property lists the operating range for the Exhalation Positive Airway Pressure (EPAP). This is the pressure setting used when you exhale. EPAP is always lower than the IPAP (inhalation) pressure, making breathing out easier and more comfortable.
A CPAP (Constant Positive Air Pressure) delivers continuous airflow at a prescribed pressure at 4cm to 30cm. APAP operates between a range and BiPAP machines are a mix of ranges and constant pressures. This pressure is measured in centimeters of water pressure (H2O) and 4-20cm is average pressure range on most CPAP machines.
This property defines whether the machine automatically adjusts pressure to meet the needs of the CPAP user on a breath-by-breath basis. All APAP machines are defined as 'Auto Adjusting', but some BiPAP's also use an Auto feature on both the IPAP and EPAP. If not marked, Yes, assume the machine does not have AUTO capabilities.
This is the maximum water capacity held by the chamber of the compatible humidifier.
This indicates that the machine has an integrated heated hose option to enhance humidification and reduce rainout. An integrated headed hose receives power directly from the machine and does not require a power supply. The tube's heat settings are set through the machine menu screen.
A yes in this field indicates the machine and humidifier have technology to reduce rainout, the water that collects in a hose when using a heated humidifier. Sensors measure the air temperature in the room and adjust the amount of heat and moisture delivered to the mask to reduce or omit rainout.
This property identifies how the machine incorporates a Heated Humidifier. Build In humidifiers are built into the CPAP machine and cannot be separated from the machine. Integrated humidifiers connect directly to the CPAP machine and they can only be used with compatible CPAP machines. With Integrated humidifiers, the CPAP machine, can be separated and used without the humidifier. CPAP machines that do not have a built in humidifier and do not have an integrated humidifier option have to use a Stand Alone humidifier as the machine's heated humidifier option. Stand Alone humidifiers can be used with any machine and connect to the CPAP machine with an additional hose.
Identifies which CPAP has a humidifier that is built into the CPAP. Built-In humidifiers cannot be separated from the CPAP machine.
This indicates which heated humidifier works best with the listed CPAP machine. In some instances, Built-in is used to describe a machine/humidifier combination system. The Fisher and Paykel HC150 was used for machines that do not have an integrated or built in humidifier system.
Measures the overall dimension of the machine only. Machines with a built-in humidifier are marked NA and must refer to Entire Size dimensions.
This is the size of the CPAP with Heated Humidifier in its travel bag.
This is the combined dimensions of the CPAP with the Heated humidifier.
This is the combined weight of the machine, heated humidifier, power cables, bag and hose. The weight of the shipping box is not included in the number.
This is the actual weight of the CPAP Machine without any accessories, power cord, or humidifier. Machines with a weight 3lbs or less are also identified in the Under 3 lbs property.
This measures only the weight of the machine and does not include total weight of power cables, bag, additional batteries or non-built-in humidifiers.
Looking to travel? We have found some machines are easier to travel with than others. These smaller and more lightweight machines are still robust enough for constant daily use as well as travel.
This indicates which manuals are included with the purchase of a CPAP machine. Most all machines come with a patient manual; also called 'user manual', which may or may not have pressure setting instructions. Some machines come with both the patient and quick setup guide or both patient and clinician manuals.
This identifies which machines have a decibel rating of 29 and lower. This does not account for use with the humidifier attached, which could supply additional volume that can muffle or amplify sound. This property does not account for noise that travels through the CPAP tubing, seemingly sounding louder than the measured db rating.
This is the level of sound the machine produces at a pressure of 10cm, standing 1 meter away or roughly 3 feet from machine. Measured in Decibels, the values also indicate whether the machine is considered a "quiet CPAP"; see Quiet property.
Warranty is the length of time that the manufacturer will guarantee a product to be free of defects. Warranties vary in length depending on manufacturer, but generally all machines come with a 2 Year Warranty.
This identifies the machine as continuous, auto adjusting, or bilevel operation. This does not identify whether the machine automatically adjusts pressure. Auto adjusting refers to APAP, whereas BiPAP Auto is differentiated using the Auto Pressure Adjustment property. BiPAP machines are defined further using Spontaneous or Timed properties.
A CPAP machine blows one constant pressure of air into your passageway throughout the night. Your ideal pressure was determined during your sleep study.
A BiPAP machine delivers two separate pressures of air; One when the patient inhales and a separate pressure when they exhale. The machine alternates between the two set pressures to help optimize your sleep apnea treatment.
An APAP machine automatically adjusts on a breath by breath basis to blow the minimum pressure needed to keep your airway open during sleep. This allows your machine to provide you with your ideal pressure nightly.
This is the company that produces the product being compared.
The Bipap Synchrony is a powerful new non invasive ventilator that combines the latest innovations of the BiPap Digital Auto-Trak Sensitivity, Vent ramp, and adjustable RiseTime technologies for more natural, and truly comfortable, ventilation. But its also more than a better BIPAP; its a revolutionary machine that can be upgraded as the users needs change.
Digital Auto-Trak Sensitivity continually monitors and adjusts to the user's breathing patterns, and along with RiseTime control allows for greater user comfort.
Compact, lightweight design weighing only six pounds,allows the unit to be easily carried. The unit is DC-compatible so it's easy to connect to an external battery.
Vent Ramp allows users to adapt to their settings by slowly increasing pressure levels to meet their prescribed pressure level in an adjustable number of breaths.
Pressure Control Mode,when integrated with other modes, offers more patient treatment options.
Oxygen Valve,when in use, aids in safe supplemental oxygen delivery.
Monitoring of Data, such as Tidal Volume and Minute Ventilation, provides users and caregivers with the knowledge of proper treatment and ventilator set-up.
This Product Includes...
Synchrony BiPAP Machine
6 Foot Hose
1 White Disposable Filter
2 Reusable Foam Filters
4cm to 30cm H2O
0 to 30 min
Starting Ramp Pressure
From 4cm up to IPAP pressure, patient-adjustable
12" x 7" x 6"
CPAP: 6 lbs
100 to 240 V, 50/60 Hz, AC Current 1.25 A Maximum; DC Voltage 12 V, DC Current 5.5 A Maximum
Maximum Heat Exposure
Machine should not be exposed to heat greater than 140 degrees Fahrenheit
CPAP.com always offers these services
24 hour access to our CPAPtalk.com Support Forum
Learning Center for information on sleep apnea and cpap equipment
Consumer Purchasing Trends Reports
Warranty Services for all items purchased through CPAP.com
Insurance Compliant Invoices
Free Prescription Request System
Philips Respironics products are for sale only to patients within the U.S. and not to other third parties or businesses.
All services provided by CPAP.com, including advertisement, ordering, sale, delivery, instruction, and setup of the products are performed solely by CPAP.com and independently of the manufacturer.
A valid prescription is required for the purchase of this product.
Digital Auto-Trak Sensitivity is a revolutionary piece of built-in software designed to detect the onset of inspiration and expiration in order to synchronize the delivery of noninvasive b-level therapy. Auto-Trak Sensitivity immediately responds to respiratory changes. assuring you the comfort they need throughout therapy. Breath-to-breath sensitivity is automatically maintained, minimizing the need for manual sensitivity adjustment. The system continually monitors and adjusts itself to accurately track the user's breathing patterns.
Digital Auto-Trak Sensitivity's advanced leak-sensing technology ensures a breath-by-breath response to your breathing patterns and circuit leaks, providing optimum support with every breath.
This machine comes with a 2 Year Manufacturer Warranty. CPAP.com will work with the manufacturer on your behalf if the machine needs repairs. Please follow these steps:
Contact us at email@example.com or 1-800-356-5221 to receive an RMA (Return Merchandise Authorization) Number before returning the equipment
Include a note with the machine describing the issue you are experiencing, along with your full contact information to include a return shipping address and the RMA number provided to you
Ship only the defective items to CPAP.com at 10101 Stafford Centre Drive, Stafford, TX 77477
Do not return accessory items such as the carry bag, hose or power cords
Email updates will be sent to keep you informed as the machine moves through the warranty process
CPAP.com will contact the manufacturer on your behalf and handle all shipping between us and the manufacturer
The repaired or replaced unit will be returned to you. Shipping charges may apply.
Please note: The manufacturer warranty is voided by water damage or misuse of the machine.
These CPAP machines operate automatically between 110v and 240v AC. When traveling either inside or outside the USA, there are no switches to switch or fuses to change. The machine will automatically sense what voltage is needed and set itself to run on that power.
The user will need to adapt the plug that goes into the wall to fit the style of outlet socket that is being used. This is simply done with a Power Adapter Plug kit. This is the same type of plug adapter that would be used with a hair dryer or an electric shaver. World Traveler Power Adapter Plugs.
This battery kit comes with an external 400 Watt DC to AC power inverter for CPAP and BiPAP machines. The kit provides AC power for machines that do not come with a DC input. Use your machine's AC cord to connect to the power inverter. The Respironics portable battery pack is an external lead acid battery that is the ultimate in power, portability and convenience. This battery kit comes with the Respironics battery and the 400 Watt DC to AC Power Inverter.
It's making a restful sleep a reality for me. It runs very less noisy than i expected, it doesn't disturb my husband at all. The alarm may need to be louder, my sleep has been so deep my husband has had to address the problem when it goes off. It is very sensitive to pressure problems. the only thing i am disappointed in is the ramp. my setting is very high, 27/21. the ramp limits allow only a short amount of time when it increases pressure.
While you may find the information provided on this site helpful, please understand that it is not in any way intended to replace the advice of a physician or medical professional. If you're concerned about your health or worried about specific symptoms, please consult a doctor. If you have additional questions about our site, please consult our User Agreement for more information.