"VERIFIED CPAP.COM CUSTOMERIt's a bit noisy, but its compact size and instant-on and instant-off features are welcome. I can breathe again!"
October 10, 2013
The PR System One REMStar 60 Series SE CPAP Machine is the entry level version of the 60 Series line of machines by Philips Respironics. Features includes Flex technology and on-board memory for compliance information. Add the OPTIONAL PR System One 60 Series Heated Tube Humidifier with Heated Tube to further enhance the comfort of the therapy with advanced humidification.
Manufactured by Philips Respironics.
The PR System One REMStar 60 Series SE CPAP Machine is an entry level single pressure CPAP machine with many features found in the 60 Series line of machines.
- Flex Technology
- OPTIONAL System One Humidification & Heated Tube
- On-board Memory for Compliance Date
- System One Resistance Control
- LED User Interface
- Travel Friendly Size
Philips Respironics offers Flex technology on the REMstar SE CPAP. It is a pressure relief system which makes the breathing process more natural and comfortable. The Flex feature of the REMstar SE senses the breathing pattern for approximately 5 breaths and bases the level of pressure relief at the time of exhalation on the breathing rate and pressure needs of the user. The Flex technology can be set to 1, 2 or 3. Or it can be turned off and there will not be a pressure drop at the time of exhalation.
Optional PR System One 60 Series Heated Tube Humidifier with Heated Tube
PR aims to maximize therapy comfort with the OPTIONAL System One Humidification System and Heated Tube. It reduces rainout and increases the level of humidification that reaches the mask. Using the room temperature in its calculations, the humidification system will operate at a System One Relative Humidity setting of 2 to deliver 80% humidity.
The LED display screen of the REMstar SE displays the 7 and 30 Day Average number of hours used. Step through the available information using the screen menu. The REMstar SE also has an on-board memory which holds up to 6 months of data. Or use an OPTIONAL generic SD card to store compliance information for up to 1 year. Visit this page to read about or order a Generic SD Memory Card. Using either the on-board or SD card to store data, the most recent data will be held in storage. To state it another way, the first data in will be the first data to fall out when capacity is reached.
System One Resistance Control
Philips Respironics includes this advanced feature which lets the machine consider the resistance of the mask to provide a consistent level of pressure. The System One Resistance Control number is stamped on mask by PR.
The LED display of the REMstar SE provides information about the machine and setting. The buttons of the machine are used to move through the various menu items relating to the operation. Information includes to:
- Pressure Setting
- Heated Humidifier
- FLEX Setting
- Ramp Starting Pressure
- Therapy Hours Time Meter
- 7 & 30 Day Average Use Time
A ramp is a comfort feature used to help ease the user into sleep. It can be set to start the pressure flow at a lower level than the prescribed therapy pressure and over a selected period of time the pressure gradually increases to the prescribed pressure. The ramp time on the SE CPAP can be set from 0 to 45 minutes and can start as low as 4 cmwp. The ramp feature is not required for CPAP therapy and can be set turned off.
Weighing just 3 lbs and only 7" x 5.5" in size, the REMstar SE CPAP machine is travel friendly. It comes with a carry case designed to accommodate the optional heated humidifier, tube and mask.
This Product Includes...
- SE CPAP Machine
- Power Supply
- Carrying Case
- 2- Reusable Foam Filters (1 Installed, 1 Additional)
- Patient Manual
Pressure Setting: Single pressure setting between 4 and 20 cm H2O (.5 increments)
Ramp Time: 0 to 45 min. (5-min. increments)
Starting Ramp Pressure: 4 to set CPAP pressure; user can adjust this setting
Altitude Settings: 1 = less than 2500 ft; 2 = 2500 to 5000 ft; 3 = 5001 to 7500 ft. Please Note: Elevations greater than 7500 ft may affect the pressure accuracy.
Dimensions - Machine Alone: 7 in L x 5.5 in W x 4 in H (18 cm L x 14 cm W x 10 cm H)
Dimensions - Machine with Heated Tube Humidifier: 10.75 in L x 6.5 in W x 4 in H (27 cm L x 16.5 cm W x 10 cm H)
Weight - Machine Alone: 2 lbs
Dimensions of 60 Watt Power Supply (For Use Without Heated Tube) - Not Including Cord or Plug: 4.75 in x 2.75 in x 1.75 in
Weight of 60 Watt Power Supply - Not Including Cord: 14.72 oz (0.92 lbs)
Dimensions of 80 Watt Power Supply (For Use With Heated Tube) - Not Including Cord or Plug: 6 in x 2.75 in x 1.75 in
Weight of 80 Watt Power Supply (For Use With Heated Tube) - Not Including Cord: 1 lb 3.04 oz
AC Power Consumption (with 60 W power supply): 100 - 240 VAC, 50/60 Hz, 2.1 A max.
AC Power Consumption (with 80 W power supply): 100 - 240 VAC, 50/60 Hz, 2.0 A max.
DC Power Consumption: 12 VDC, 6.67 A
Sound Level : Less than 30 dBA
22mm Performance Tube Length: 72 inches
Optional Heated Hose Length: 72 inches
Filters: Pollen (Foam) and optional ultra-fine (Disposable)
Device Set-Up: LED Display & Device Keypad
- Data Type: Compliance
- Compliance: VIC 7-30 Day Averages
- SD Card: Up to 1 year of Data
- On-board Memory: up to 6 months
Electrical Requirements: 100 - 240 VAC, 50/60 Hz, 12volt DC
European Voltage Usage: Yes, outlet adapters required
Max Exposure Heat for Machine: Machine should not be exposed to heat greater than 140 degrees Fahrenheit
Optional Heated Tubing Specifications
Maximum Recommended Pressure: 25 cm H20
Inner Diameter: 15 mm (0.6 in.)
Length: 1.83 m (6 ft)
Heated Tubing Temperature Range: 16º to 32º C (60º to 89º)
Heated Tubing Temperature Cut-out: ? 41º C (? 106º F)
Material: Flexible plastic and electrical components
Required Settings: 1 Pressure Setting is Required
Unless directed otherwise, CPAP.com will leave this machine at its default settings of:
- C-flex: 2
- Ramp: On
- Ramp Time: 5 Minutes
- Ramp Start: 4 cm H2O
CPAP.com always offers these services:
- 8 am to 8 pm CST Toll Free Phone and Email Support
- 24 hour access to our CPAPtalk.com Support Forum
- Learning Center for information on sleep apnea and cpap equipment
- Consumer Purchasing Trends Reports
- Warranty Services for all items purchased through CPAP.com
- Insurance Compliant Invoices
- Free Prescription Request System
- A valid prescription is required for the purchase of this product.
- Philips Respironics products are for sale only to patients within the U.S. and not to other third parties or businesses.
- All services provided by CPAP.com, including advertisement, ordering, sale, delivery, instruction, and setup of the products are performed solely by CPAP.com and independently of the manufacturer.
Product Name: The SE in this product names stands for "Standard Edition".
On Board Memory: The on-board memory of the REMstar SE holds up to 6 months of data. If an SD card (optional, not included with purchase) has been inserted into the slot, it will hold up to 12 months of data. To view or purchase an SD card visit this page: SD Memory Card for IntelliPAP, PR System One and S9 Machines.This machine is NOT compatible with the SleepMapper application.
Backlight Setting: The backlight on the SE can be changed to suit the user's needs. In the patient menu screen select the icon that resembles a glowing light bulb. Using the right & left arrow keys select 0 (OFF) or 1, 2 or 3 for the level of brightness.
Flex Setting: The Flex setting on the SE is represented by the word "FLEX" in the patient menu screen. It can be set to 0 (OFF); 1, 2 or 3. The Flex setting of 1 provides the smallest amount of pressure relief while 3 provides the greatest relief.
Power Supply and Cord Included
Philips Respironics includes the power supply and cord with the purchase of this machine. Find a black flap secured with a hook and loop fastener on the interior wall of the bag. By pulling on the fabric tab, the fastener will release. The power supply and cord are placed within the pocket.
The PR System One Heated Tube for 60 Series Machines is a heated hose that integrates with the 60 Series Heated Humidifier. The heated tube connects to electrical ports on the humidifier.
Not using the Heated Humidifier? The heated tube can still be used by connecting to ports on the air outlet on the side of the machine. In this configuration, the hose will not heat and will act as a standard hose.
System One Resistance Control Settings
Please reference this link to find the resistance control setting for the different Philips Respironics masks: System One Resistance Control Settings.
Tube Type Setting: The SE is able to accommodate either a standard 22mm (22) tube or the slim 15mm (15) tube. This setting is changed in the patient menu screen by selecting the icon that looks like a tube. If the SE is being used with the OPTIONAL Heated Tubing & Heated Humidifier, the machine will AUTOMATICALLY select the appropriate setting (15 h) which is a 15mm heated tube. The tube setting cannot be changed if the heated tube is being used.
Please Note: If the 60 Series SE CPAP machine is purchased individually it will include a 60 Watt power supply. If the 60 Series SE CPAP machine is purchased with the heated tube humidifier and heated tube, it will include an 80 Watt power supply. The 60 Watt power supply will not power the heated tube. If using the Heated Tube Humidifier and Heated Tube, the 80 Watt power supply must be used to successfully heat the tube. The 60 watt supply weighs 10.95 oz and is 4.5" L x 2.75" W x 3" H with the cord wrapped around. The 80 watt supply weighs 1 lb 3 oz and is 6" L x 2.75" W x 3.25" H with the cord wrapped around.
The 60 Series SE machine comes with a 2 Year Manufacturer Warranty. CPAP.com will work with the manufacturer on your behalf if the machine needs repairs. Please follow these steps:
- Contact us at email@example.com or 1-800-356-5221 to receive an RMA (Return Merchandise Authorization) Number before returning the equipment
- Include a note with the machine describing the issue you are experiencing, along with your full contact information to include a return shipping address and the RMA number provided to you
- Ship only the defective items to CPAP.com at 13235 N Promenade Blvd Stafford, TX 77477
- Do not return accessory items such as the carry bag, hose or power cords
- Loaner Machines are available for rental during the warranty process
- Email updates will be sent to keep you informed as the machine moves through the warranty process
- CPAP.com will contact the manufacturer on your behalf and handle all shipping between us and the manufacturer
- The repaired or replaced unit will be returned to you. Shipping charges may apply.
Please note: The manufacturer warranty is voided by water damage or misuse of the machine.
This system operates automatically between 110v and 240v AC. When traveling either inside or outside the USA, there are no switches to switch or fuses to change. The machine will automatically sense what voltage is needed and set itself to run on that power.
When traveling outside the US, the user will need to adapt the plug that goes into the wall to fit the style of outlet socket that is being used. This is simply done with a World Traveler Power Adapter Plug kit. This is the same type of plug adapter that would be used with a hair dryer or an electric shaver.
Optional DC Cable
The machine can operate using DC Power using the Shielded DC Cord for PR System One 60 Series Machines. This cord is compatible ONLY with PR System One 60 Series Machines. The 12 Volt DC Power Cord (connects CPAP to cigarette lighter socket) used with other PR System One machines IS NOT compatible with PR System One 60 Series Machines.
Weighing in at only 1.16 pounds the C-100 Battery is a terrific option for travel. Plus, the C-100 comes with a DC cable and connector peg suitable for use with the PR System One REMStar 60 Series SE CPAP Machine, so no need to purchase a separate DC cable. This lithium ion battery meets UN Manual Test and Criteria requirements for travel. The C-100 Pack includes many accessories and a handy carry case designed to allow the the cords to be attached with the battery inside the case.
This portable battery pack is an external lead acid battery that is the ultimate in power, portability and convenience. This battery kit comes with the Respironics Battery Pack and the Shielded DC Cord for PR System One 60 Series Machines.