Respironics Battery Pack (No DC Cable)
The Respironics Battery Pack (No DC Cable) is the universal portable battery pack from Respironics' line of DC accessories. It provides the ultimate in power, portability and convenience. This is the battery pack, pouch and cord and does not include a DC cable to power a machine.
Manufactured by Philips Respironics.
4.8 of 5 72 Reviews
Available to Ship: Tomorrow
Have a question? Call us!1-800-356-5221(Mon-Fri 8AM-10PM CST, Sat 8AM-6PM CST)
Sealed Lead-acid Battery: Respironics uses a 14.4 Amp Hour sealed, lead-acid battery. The battery comes with a convenient carry case to store the battery. The charger is also included.
Switch Mode Charger: The 12V, 2.3A charger is designed to effectively charge a sealed, lead-acid battery while protecting the system from overcharging. NOTE: Remove the charger from the carry case PRIOR to charging the battery. Do not charge the battery with the charger in the carry case. The Respironics Battery Pack must be charged from an AC source.
Universal Voltage Input of 100-240 AC: The Respironics Battery Pack can be charged with AC current between 100 - 240V. This provides reverse polarity and short circuit protection.
Status Lights: Multi-colored charging status lights:
Charge Time: Charging the battery from 0 (zero) to a full charge takes approximately 10 hours. If the battery has residual power stored, the charge time will be shorter. The battery MUST BE in the upright position during charging and use. Remove the battery from the charger once fully charged. Fully charge the battery before use. The battery should be fully charged before being stored. See the Specs Tab of this product page for more information on storage and usage.
Shelf Life: The Respironics Battery Pack has a charge shelf life of 6 - 9 months. Charge the battery fully prior to storage. The optimal temperature for storage 77°F. As the battery sits there will be a self discharge of power, with less residual charge as time passes. Do not allow the charge level to go below 20% - 25%. See the Specs Tab of this product page for more information on self discharge rates.
Estimated Run Times: Please see the "Machine Pressure and Battery Performance" feature below to estimate the battery run time for your machine at your pressure setting. Using a DC capable heated humidifier on a 12 volt battery power source will significantly reduce the amount of power provided. In most cases, battery run time is reduced to less than half the estimated operational time. Some older heated humidifiers will not heat with a DC power source and will instead act as a cool passover humidifier. CPAPs or BiPAPs requiring inverters reduce the power time even more.
12v DC Cable NOT Included: This product does not include a 12 DC cable needed to power a machine: This product - Respironics Battery Pack (No DC Cable) - is perfect for a user who already has a 12v DC cable for their CPAP machine.
Complete Battery Kits Available: If you do not have a DC cable for your machine, you can select from one of the Battery Kits below that come with the parts needed to run the Respironics battery with your machine. Each Battery Kit lists the machines that are compatible for that kit. If you have any questions call our toll free number 800-356-5221 for assistance from a CPAP expert.
*Note: This is a battery life estimation based on the battery and the machine. Each machine has unique power needs. Look for the Battery kit that has all the cables needed to safely run your machine. Each battery kit page contains a note about which machines are compatible with that specific battery kit. Please always read the "Important Tips" tab for further information on battery use.