PR System One REMStar 60 Series Pro CPAP Machine with Bluetooth
Sorry, this item is no longer available.
Check out the newer model hereDreamStation Pro CPAP Machine
Remstar 60 Pro CPAP w/Bluetooth Overview
The PR System One REMStar 60 Series Pro with Bluetooth is designed to relay sleep results to the user by connecting to Respironics' SleepMapper app. Download SleepMapper on an apple or android smartphone or tablet for easy accessibility. For increased comfort during therapy, connect the machine to the OPTIONAL integrated humidifier to allow warm air to travel through the heated tube into the CPAP mask.
Philips Respironics 60 Series Pro with Bluetooth uses C-Flex Plus to soften the transition between inhalation and exhalation to help the user adjust to CPAP pressure. Auto-Trial mode and CPAP-Check mode are additional features used to provide pressure relief. Pair the 60 Series Pro using Bluetooth to the SleepMapper app to review therapy results the next morning.
- Bluetooth Connection to SleepMapper
- OPTIONAL Heated Tube Humidifier
- System One Dry Box
- Smart Technology Modes
- C-Flex Plus Pressure Relief
- System One Resistance Control
Bluetooth Connection to SleepMapper
Bluetooth wireless technology has been included in the design of the machine to allow users to easily access their sleep results. Download the SleepMapper app onto a smartphone or tablet to direct the data to your account. Once installed, the app will pair with the machine each night to display results the next morning. The results may be used to review compliance between a doctor and patient.
OPTIONAL Heated Tube Humidifier
The Bluetooth 60 Series Pro CPAP machine attaches to the optional integrated humidifier to provide warm air from the CPAP machine to the mask. Attach the optional heated tube humidifier to add moisture from the water chamber into the therapy air. The optional heated tube included with the humidifier connects to the humidifier port so the warm air can maintain heat as it travels through the hose. Heating the air as it travels from the CPAP machine may diminish rainout and moisture in the mask.
System One Dry Box
The System One Dry Box technology reduces the chance of water damage to the 60 Series CPAP machine. The center air outlet port uses unique design features to help prevent water from exiting the chamber and entering the machine. The center air outlet port uses a one way valve. This location on the top of the chamber helps to minimize total water displacement. If water were to escape the chamber it is trapped by the Dry Box.
Smart Technology Modes
The smart technology of Auto-Trial Mode and CPAP-Check Mode offer up to 30 nights of auto adjusting capability. During that period of use the system functions as an Auto CPAP delivering an adaptive pressure on a breath by breath basis. Following the set time of the Auto-Trial mode, the machine functions as a CPAP delivering one set pressure that the patient was at or below 90% of the time.
The CPAP-Check mode continuously verifies pressure requirements and makes adjustments after every 30 hours of use providing the user with more effective pressure levels over time. An Auto-Trial mode must be a minimum of 3 nights in length. The entire available 30 nights of Auto-Trial do not need to be used continuously. Visit the "Important Tips" tab of this product page to read an example of the Auto-Trial and CPAP-Check modes.
PLEASE NOTE: This is NOT an Auto CPAP Machine. It functions as an Auto CPAP during Auto-Trial Mode and switches to CPAP mode during CPAP-Check mode.
C-Flex Plus Pressure Relief
Breathing naturally during CPAP therapy is accomplished through features such as C-Flex Plus, which are designed to soften CPAP pressure as the user transitions between inhalation and exhalation. Philips Respironics was a pioneer in offering machines with pressure relief, and they continue to use these features in the PR 60 Series Pro with Bluetooth.
System One Resistance Control
System One Resistance Control was developed by Respironics to help their machines reach the optimal pressure by using a numbering system which allows the CPAP device to recognize the mask being used with treatment. It achieves accuracy and reliability by instructing the device to compensate for the specific mask type. Please Note: Resipronics' masks are the only masks designed with a resistance control number. When planning to use this machine feature, be sure to use a Respironics mask.
This Product Includes...
- 60 Series Pro CPAP Machine with Bluetooth
- Power Supply and Cord
- Carrying Case
- SD Card
- Reusable Foam Filter (2)
- Patient Manual
- Clinical Quick Setup Guide
Dimensions of 60 Watt Power Supply (For Use WITHOUT Heated Tube) - Not Including Cord or Plug: 4.75 in x 2.75 in x 1.75 in
Weight of 60 Watt Power Supply - Not Including Cord: 14.72 oz (0.92 lbs)
- Display: Compliance VIC. 7-Day and 30-Day Averages
- SD Card > 1 Year Compliance Data
- On-Board: Up to 6 Months
- C-Flex: 2
- Ramp: On
- Ramp Time: 5 Minutes
- Ramp Start: 4 cm H2O
- Auto On: On
- Auto Off: Off
CPAP.com always offers these services
- 24 hour access to our CPAPtalk.com Support Forum
- Learning Center for information on sleep apnea and cpap equipment
- Consumer Purchasing Trends Reports
- Warranty Services for all items purchased through CPAP.com
- Insurance Compliant Invoices
- Free Prescription Request System
- Philips Respironics products are for sale only to patients within the U.S. and not to other third parties or businesses.
- All services provided by CPAP.com, including advertisement, ordering, sale, delivery, instruction, and setup of the products are performed solely by CPAP.com and independently of the manufacturer.
- A valid prescription is required for the purchase of this product.
Pairing Bluetooth with Mobile Devices
Bluetooth technology allows for a wireless transfer of data from the machine to the designated SleepMapper app to deliver sleep results to a handheld device.
SleepMapper Mobile Compatibility
The SleepMapper app may be used with an iPhone iOS 6 device or newer as well as an Android 2.2 OS or newer. Please Note: A smartphone is not included with the purchase of the 60 Series Pro with Bluetooth.
SleepMapper App Tips
- SleepMapper will only work with Philips Respironics machines.
- Install the SleepMapper App on your phone or tablet for free.
- Visit the i-Tunes store or the Google Play website to download the SleepMapper app.
- The machine SD card can be used in conjunction with SleepMapper.
SleepMapper Name Changed to DreamMapper
When searching for the Respironics' SleepMapper app on an Apple or Android device, the name SleepMapper will no longer appear in iTunes or Google Play. Instead, the web based app displays the name DreamMapper, which is the newer version of the app. Please Note: Though the name has changed, the information provided remains the same. DreamMapper is still compatible with all 60 series machines which use the bluetooth feature.
Machine and Humidifier Compatibility
- System One 60 Series Heated Tube Humidifier with Heated Tube can be used ONLY with PR System One REMStar 60 Series machines.
- PR System One Heated Humidifiers (those that do not include the words "60 Series" in the name) cannot be used with PR System One 60 Series Machines.
- The 60 Series heated humidifiers CANNOT be used with any other machines made by Philips Respironics. This includes other versions of the PR System, One REMStar line of machines.
- The 60 Series heated humidifiers are compatible only with PR System One REMStar 60 Series machines.
Side Cover Panel
PLEASE NOTE: The manufacturer does NOT include the side cover panel when a 60 Series machine and heated humidifier are purchased together as a set. This part may be purchased separately:Side Panel for PR System One 60 Series CPAP Machines.
Power Supply and Cord Included
If the 60 Series machine alone or machine and standard (non-heated tube) humidifier are purchased, the External 60 Watt Power Supply for PR System One REMstar 60 Series Machines and a power cord will be included.
If the CPAP Pro is purchased with the heated tube humidifier, an 80 Watt Power Supply and power cord are included. The DreamStation 80 Watt Power Supply is the newer version and is compatible with 60 Series machines.
The power supply and cord are found in a pocket behind the black flap on the interior wall of the travel bag. Pull the fabric tab to release the fastener and access the equipment.
PR System One Humidifier Modes
PR System One REMStar 60 Series machines version 1.22 or higher have two humidification modes:
- Classic Humidification
- System One Humidification
Version Note: To determine the version of the software installed on a PR System One REMStar 60 machine, unplug the machine and plug it back into an outlet. The version is displayed on the screen when the unit is first plugged into an outlet. The humidification settings are displayed only when a PR System One Heated Humidifier is attached to the machine.
Classic Mode Humidification
The Classic mode of humidification allows the CPAP user to control the heat setting on the humidifier. The user will select "Classic" mode settings C1, C2, C3, C4, or C5. The "C" in front of the humidification setting indicates "Classic." C1 is the lowest level of humidification and C5 is the highest level of humidification.
Please Note: If the PR System One Heated Tube for 60 Series Machines is attached to the humidifier the Classic mode humidification will not be offered, only System One Humidification Mode. To use the Classic mode of humidification attach a non-heated tube to the humidifier.
System One Humidification Mode
The System One Humidification uses the technology of the system to set the level of humidification delivered. The machine makes the decision based on the humidity present in the room air. The CPAP user decides how much they want to receive. A setting of 1 delivers the lowest humidity level and a setting of 5 delivers the highest level. The humidification setting will show 1 - 5. The letter "C" will NOT precede the number.
Passover Humidification, Heat Off
To turn the heated humidifier off, set the humidifier to zero "0". The air from the machine will pass over the water in the chamber acting as a passover Humidifier.
Setting the Humidification Mode and LevelPlease Note: The humidifier menu will only show when the humidifier is attached to the machine.
First, use the wheel on the machine to select the 'Setup' menu. Then, scroll through the menu to locate SYSTEM ONE Humidification. Select 'On' to allow the machine to determine the level of humidification delivered throughout the system based on the 1-5 setting. When in the SYSTEM ONE Humidification mode there is no letter indicator next to the number.
To control heat manually, select the 'Off' option. Then, set the humidification level by adjusting the number to the right of 'Humidifier,' which is found under SYSTEM ONE Humidification. Select a number within the range of 1 to 5. The letter "C" found in front of the humidification number represents the word 'Classic'. The lowest heat is recorded as C1, while C5 is the highest heat level.
Power Supply and DC Cable Compatibility Tips
Please Note: To power the humidifier, the Respironics' 80 Watt power supply is included with the purchase of a 60 Series Pro with Bluetooth and Heated Tube Humidifier. If the user chooses to use the CPAP machine without the humidifier and heated tube, the 60 Watt power supply may be used.
The External 80 Watt Power Supply for PR System One REMstar 60 Series Machines is NOT compatible with other versions of PR System One machines. The connection peg that plugs into a PR System One REMStar 60 Series machine is a specific size and will fit only this machine. The external power supply for this machine has different specifications than the External Power Supply for PR System One REMstar CPAP and BiPAP Machines used with other machines in the Philips Respironics lines. The Power Cord that plugs the external power supply into the wall outlet is the same for all System One Machines.
The Shielded DC Cord for PR System One 60 Series Machines that is compatible with PR System One 60 Series Machines works ONLY with PR System One 60 Series Machines. The 12 Volt DC Power Cord (connects CPAP to cigarette lighter socket) used with other PR System One machines IS NOT compatible with 60 Series Machines.
System One Resistance Control Settings
Please reference this link to find the resistance control setting for the different Philips Respironics masks: System One Resistance Control Settings.
The PR System One Heated Tube for 60 Series Machines is included with the purchase of this CPAP machine and humidifier. The heated hose connects to the electrical ports on the 60 Series Heated Humidifier to keep the air warm as it travels from the humidifier through the hose to the CPAP mask.
Not using the Heated Humidifier? The heated tube can connect to the ports on the air outlet on the side of the machine. In this configuration, the hose will not create heat and will act as a standard hose.
In addition to the 60 Series heated tube, a traditional CPAP hose can be used. Set the size of hose you will be using in the Set Up Menu under "Tubing Type". Select either 15 mm or 22 mm depending on the size tube being used.
Identifying PR System One REMStar 60 Series MachinesThe following highlight physical characteristics, which identify a PR System One REMStar 60 Series Machine:
System One 60 Series Heated Tube Set-Up
Philips Respironics provides these instructions for setting up the heated tube: System One 60 Series Heated Tube Set-Up
Example of Auto-Trial and CPAP-Check Modes:
In this example, the PR System One 60 Series Pro is set in Auto-Trial mode for 5 nights of use with an ideal pressure of 10cm H2O. During this time the machine functions as an auto-titrating machine, adjusting the pressure on a breath by breath basis, all night long. At the end of 5 nights of use the Auto-Trial mode ends. Now, the machine moves into the CPAP-Check mode of operation delivering a pressure of 10cm H2O. After 30 hours of use the CPAP-Check mode data indicates the pressure needs to be raised. The machine will automatically change the pressure to 11cm H2O.
Alternately, if the pressure needs to be lowered, the machine will automatically change to 9cm H2O. If no pressure adjustment is needed, the pressure will stay at 10cm H2O. The CPAP-Check mode will continue every 30 hours to monitor and adjust the pressure as needed. Please Note: The pressure will not raise or lower more than 3cm H2O from the original pressure. Using the original pressure of 10cm H2O, the machine will not drop the pressure below 7cm H2O or raise the pressure above 13cm H2O.
If the machine is placed back into Auto-Trial mode mode to determine a new ideal pressure, the CPAP-Check mode would monitor and change pressure as needed every 30 hours, staying within 3 cm/H2O of the new ideal pressure.
The machine does not need to be set on CPAP-Check as it will automatically move into the mode following Auto-Trial. It can be placed on straight CPAP mode, so no pressure checks or adjustments are made.
What is the Respiratory Distress Index? The RDI is used to determine the severity of sleep apnea. The index value is determined based on the number of apneas, hypopneas, and respiratory event-related arousal (RERAs) that occur per hour of sleep.
This system operates automatically between 110v and 240v AC. When traveling either inside or outside the USA, there are no switches to switch or fuses to change. The machine will automatically sense what voltage is needed and set itself to run on that power.
When traveling outside the US, the user will need to adapt the plug that goes into the wall to fit the style of outlet socket that is being used. This is simply done with a World Traveler Power Adapter Plug kit. This is the same type of plug adapter that would be used with a hair dryer or an electric shaver.
Optional DC Cable
The machine can operate using DC Power using the Shielded DC Cord for PR System One 60 Series Machines. This cord is compatible ONLY with PR System One 60 Series Machines. The 12 Volt DC Power Cord (connects CPAP to cigarette lighter socket) used with other PR System One machines IS NOT compatible with PR System One 60 Series Machines.
Weighing in at only 1.7 pounds the Freedom Battery is a terrific option for travel. Plus, the Freedom comes with a DC cable and connector peg suitable for use with the PR System One 60 Series Pro, so no need to purchase a separate DC cable. This lithium ion battery meets UN Manual Test and Criteria requirements for travel.
This portable battery pack is an external lead acid battery that is the ultimate in power, portability and convenience. This battery kit comes with the Respironics Battery Pack and the Shielded DC Cord for PR System One 60 Series Machines.
The Respironics Lithium Ion Travel Battery Kit is intended for use with 50 Series, 60 Series, and DreamStation machines. Uninterruptible Power Supply (UPS) mode is designed to help the battery kick in automatically if AC power fails. This mode ensures power is never lost to the CPAP machine, even after power from the outlet is disconnected. When AC power is available again, the battery will be bypassed, and AC will resume powering the machine. The battery weighs 2.25 lbs and is FAA approved for use on flights. The kit includes a travel case, device cord, and 50 Series adapters used to connect the machine to the battery as well as connect the battery to the power supply for charging. This kit does not include a power supply as it is designed to operate from the machine's power supply.
The manufacturer states the Respironics battery has a run time of 14 hours with a pressure of 10cm H20 or lower, excluding a heated tube humidifier.
Just a tad noiser compared to more modern machines but at the time the price was unbeatable. A good machine overall. more...
I have used CPAP.com for years now and I have always been pleased with their pricing and service. I have found most of their products to be priced low and their service is always excellent. Their website is easy to use and transaction turn-a-round is speedy. Highly recommend this site as your CPAP Supplier. more...
Got this unit to replace my 12 year old Resmed CPAP. Much quieter than my old unit, and many more options to make it's use much more comfortable than my old one. Hopefully this one will last 12+ years, too! more...
Amazining advanced caring machine runs extreme quiet and efficiently providing information necessary to understanding present therapy and set goals for successful health management more...
While this model is being closed out it is a great deal and a better CPAP then my previous System One machine from a local supplier. With a pressure setting of 9.5 cm, I really appreciate the C-Flex Plus as it make it much easier to exhale. I also am enjoying the quality of sleep charting I get using the Bluetooth connection with compatible DreamMapper application (not SleepMapper). more...
Compare Similar Products
Basic FeaturesManufacturerPhilips RespironicsResMedApex MedicalTypeCPAPCPAPCPAPWarranty2 Years2 Years2 YearsSound Level<30 dBA26.6 dBA28 dBQuietYesYesYesManuals IncludedPatientPatient/Quick SetupPatient
DimensionsTravel SizedYesYesYesUnder 3 lbsYesYesYesMachine Weight2.3 lbs2.751.87 lbsEntire Weight6.3 lbs5.1 lbs5.4 lbsEntire Size6.5" x 10.75" x 4"10.04"x4.75"x5.91"10 in x 6.75 in x 5.5 inEntire Package Size8 " x 14.75" x 8"13.24"x10"x5"13 in x 7.5 in x 8.5 inMachine Size Only7" x 5.5" x 4"NA5.7 in x 5.1 in x 3.9 in
Humidifier FeaturesHeated HumidifierSystem One 60 SeriesHumidAirXT Heated HumidifierBuilt-In HumidifierNoYesNoHeated Humidifier TypeIntegratedBuilt InIntegratedAmbient Rainout Reduction60 Series Heated TubeAutoClimateControlNoIntegrated Heated HoseYesClimateLineAireNoHumidifier Water Capacity325 mL380mL300 mL
Pressure FeaturesAuto Pressure AdjustmentNoNoNoPressure Range4-20 cm H2O4-20cm H2O4-20cm H2OEPAP Pressure RangeNANANAEasy BreathingC-Flex PlusEPR with Easy-BreatheNoRampYesYesYesCPAP ModeYesYesYesLeak CompensationYesYesNoMax Operating Altitude7,500 ft8,500 ft8,000 ft (4-18 cm)
Power FeaturesDirect Battery OperationYesYesNoVoltage Range100-240V AC100-240V100-240V ACOptional DC CableYesYesNoIntegrated BatteryNoNoNo
Software FeaturesAdvanced Software/DataMSDYesNoData CardYesYesNoOptional SoftwareEncoreViewerResScanNo
Additional FeaturesAuto Altitude AdjustmentAutoAutoManualAuto ON/OFFYesYesNoMask Off AlertLCD; ALARMNoNoSpontaneousNoNoNoTimedNoNoNoDesigned For HerNoNoNoBack Lit DisplayYesYesYesBacklight Shuts OffYesYesNo